# Setting Examples - Sharp IV-S30 User Manual

Compact image sensor camera function and operation

Numerical Calculations

## 15-3 Setting examples

The example below shows the settings used to evaluate and output the X-Y coordinates (differences)
of the detection points in images 0 and 1, using the evaluation calculation. (Degree of match
inspection - See Chapter 5; Output - See Chapter 16: PC Function.)
Point detected in image 0
(register number 00, model 0)
(1) Specify the images shown above in the conditions for a degree of match inspection
- After specifying the images, return to the [TYPE MEAS COND] menu for the degree of match inspection.
(2) Operations on the [TYPE MEAS COND] menu
[TYPE MEAS COND]
6MEASUREMENT 1
[MEAS COND]
1DTECT PRECISION
2REGISTER NO.
3SEARCH
4MATCHING
REGISTER NO.
5MDL0
6MDL1
NUMERIC CALC
1
OPS-MENU RETURN LOCK EVALUATION DST&ANG NUM-CALC OUT-COND
1. Move the cursor to the "NUM-CALC" item, and press the SET key.
- The [NUMERIC CALC] menu will be displayed.
(3) Operations on the [NUMERIC CALC] menu
[NUMERIC CALC]
1CALC RESULT N00(0~15) NO YES
2TYPE
1
2
3FORMULA
3
4FUNCTION
15
5UPR&LOW LIMIT +0000000.0000~+0000000.0000
6OUTPUT
4
7RUN A TEST
6
N00
7
N01
N02
N03
1. Move the cursor to the "1CALC RESULT" item using the up and down keys, and press the SET key. Select
"N00" using the up and down keys, and move the cursor to "YES" using the left and right keys. Then press
the SET key.
- Items 2 to 7 will be displayed.
2. Move the cursor to the "2TYPE" item using the up and down keys, and press the SET key. Then move the
To the next page
Image 0
Y coordinate difference
CHK-DEG-MATCH C1
STANDARD HIGH
00(0~15) NO 1P-SCH 2P-SCH
YES NO
AVG-GRAY DIFF.ABS
00 01 02 03 04 05 06 07 08 09 10 11 12 13 14 15
S S
MATCH[M] COORD[X Y] LVL[G]
NUM-CAL[NC] CNST[C]
REG0(0~15) MDL0(0~1) N00(0~15)
CNST0000.0 + -
/
DEL END
*
SUM AVG REG00~00(0~15)
NO Y0(0~7) C000(0~107)
(SET KEY)
X coordinate difference
Object to be measured
Point detected in image 1
(register number 01, model 0)
Image 1
(MEAS-COND)
15-12

#### This manual is also suitable for:

Iv-s31mIv-s32mIv-s33m