Chapter 17: Setting the Input/Output Conditions
This chapter explains the input/output settings on the IV-S30. The conditions are set on the [I/O
 When you want to select "PARALLEL + SERIAL + USB" on the "MEAS INP I/F"
On the MAIN OPS MENU, move the cursor to "SYS-CND," and press the SET key.
-On the [SYSTEM COND] menu, move the cursor to item "1I/O CONDITIONS"
and press the SET key.
- On the [I/O CONDITONS] menu, select "1 MEAS INP I/F" and press the SET key.
Then, select "PARALLEL + SERIAL + USB", and press the SET key.
1I/O CONDITIONS (TO NEXT SUB-MENU)
1MEAS INP I/F
2SERIAL OUTPUT NO PC-LINK SERIAL
3MANL TYPE CHNG NO YES
4PARALLEL INP X6 64OBJ-CHG REG REF(MEAS0 COMPARE-IMGS)
5PARALLEL INP X7 EXT-INP CHNG-IMG-OUT-CAM CAM-MEAS 2 IMAGES
OPS-MENU RETURN LOCK
1 MEAS INP I/F
(measurement start input)
2 SERIAL OUTPUT
3 MANL TYPE CHNG
4 PARALLEL INP X6
To the next page
(TO NEXT SUB-MENU)
CAPTURE COMPLETE MEAS-COMPLETE
Select an input interface for starting the measurement on the IV-S30.
Select "PC-LINK (computer link)", "SERIAL (general-purpose serial)", or "NO."
The object type can be changed manually using the remote keypad on the MAIN
Æ YES: Means that you will be allowed to change the object type manually on the
MAIN OPS MENU. Move the cursor to the "CHG-TYPE" item on the MAIN
OPS MENU, and select the object type No., displayed on the upper part of
the screen, using the up and down keys.
Æ NO: Means that you cannot change the object type manually on the MAIN
Select the type of input terminal (INPUT) X6.
Æ 64OBJ-CHG (Change 64 object types)
Select when you want to change the object type number (0 to 63) using the
Æ REG REF (MEAS 0 to 4)
- When X6 is turned from OFF to ON, and gray search is selected, the
reference image will be registered in the SDRAM. When edge detection is
selected, the reference coordinates will also be registered.
- To store the reference image in flash memory
: Select the SAVE key on any sub menu or use a general purpose serial
- Available measurement programs (gray scale search)
: Positional deviation measurement, the degree of match inspection, multiple
positions measurement, multiple degree of match inspections for register No. 0,
- Applicable measurement program (edge detection)
: Positional deviation measurement
- If a reference image has not been specified, an "UNABLE REGISTER
REF.IMG" (X6 reference image register error (code3E)) will occur.
Setting the Input/Output Conditions
Only displayed here when
the IV-S33M is used.
(When the IV-S31M/S32M
is used, "EXT-INP" will be
After selecting "REG-
REF," chang 0 to 4 using
up and down keys.
Change Y0 to Y7 using
the up and down keys.