# Final Numerical Calculations - Sharp IV-S30 User Manual

Compact image sensor camera function and operation

Numerical Calculations

## 15-2 Final numerical calculations

Final numerical calculations can be set at item "FINAL-CALC" on the conditions of object types menu
[TYPE MEAS COND].
The setting procedure, error output settings, calculation sequence and parallel output based on the
output conditions are exactly the same as those used for item "15-1 The individual numerical
calculations for each measuring program."
On the [TYPE MEAS COND] menu, move the cursor to "FINAL-CALC (final numeric
calculation)" and press the SET key.
[FINAL NUM.CALC]
1CALC.RESULT AN00(0~15) NO YES
2OBJECT TYPE MEAS-CAL-RESULT[N] A-CAL-RESULT[AN]
3FORMULA
4FUNCTION
5UPR&LOW LIMIT
6OUTPUT
7RUN A TEST
AN00
AN01
AN02
AN03
- The display areas on the screen are in the same places as those for item
individual numerical calculations for each measuring program.
1 CALC.RESULT:
2 OBJECT TYPE:
3 FORMULA:
(registaration number)
MEAS-CAL-RESULT [N]
A-CAL-RESULT [AN]
15
NUM-CAL [NC]
CNST [C]
Note: Please use a smaller number for entering numbers at AN (00 to 15) than that 1
4 FUNCTION
5 UPR&LOW LIMIT:
6 OUTPUT:
7 RUN A TEST:
NUM-CAL[NC] CNST[C]
REG000(0~127)
CNST0000.0 + -
*
SUM AVG AN00~00(0~15)
0000000.0000~
+
NO Y0(0~7) C000(0~107)
(SET KEY)
N 01N00+02N00+3N01
+0009800.0000~+0010000.0000 +00009800.0000
The location where the rule 4 calculation result of the specified output data is
stored may be set from 00 to 15.
Select the type of data being calculated.
A number of style settings are used.
The first line will change according to the selection made in item 2.
Selection on line 2
MEAS0 (0 to 4) CAM1 (1 to 2) N00 (0 to 15)
CNST0000.0 + -
AN00 (0 to 15)
CNST0000.0 + -
ABS SQRT TAN ATAN AN00 (0 to 14)
+0000000.00 (the second line will not be displayed)
CALC. RESULT (calculations results) AN (00 to 15).
When you want to total (SUM) or average (AVE) the calculation results AN,
specify the range (0 to 15).
Enter the upper and lower limits for making a decision.
Setting the output of the calculation results.
Pressing the SET key will record the setting details as well as run a test.
N00(0~15)
/
DEL END
0000000.0000
+
The first line of the display
/
DEL END
/
DEL END
MAX MIN AN00 to 00 (0 to 14)
15-10
Example:
"Results of the calcula-
tions for measurement 0,
camera 1: N00" +
"Results of the calcula-
tions for measurement 0,
Y1
camera 2: N00" +
OK
"Results of the calcula-
tions for measurement 3:
N01"
15-1 The
"
"