Continued from the previous page
(6) Degree of match inspection
Press the TRG/BRT key, and the IV-S30 will calculate the X-Y coordinates of the detection points in
images 1 and 2, and then evaluate and output the results.
MEAS XXXXms XXXX-XX-XX XX:XX
MEASUREMENT 1 CHK-DEG-MATCH
CHG-MEA CHG-REG CHG-RST CHG-C1
OPS-CND MEA-CND SYS-CND CHG-TYPE NEXT-NG RE-EXAM-NG MANL-MEAS
Final output conditions
* Move the cursor to the "CHG-RST" item, and press the up or down key. Then, the numeric calculation results
will be displayed.
CHG-C2 CHG-EVAL CUSTOM-MNU
If the difference in the X and Y coordinates is within the range set on the numeric
calculation menu, Y0 will be turned ON, and a filled box
difference is not within the range, Y0 will be turned OFF, and an empty box
F.IMG C1 L
- If the result is within the range,
N00 and N01 will be turned ON
(OK), the auxiliary relay C000 will
be turned ON, and output Y00 will
be turned ON.
will be displayed. If the