Setting and Operating Outline
1-7 Remote key pad and specifications
(up, down, left
Set (select) key
* The direction keys have an auto-repeat function
1 Direction keys
left and right)
Selecting an item on a menu
Setting a window
Setting a value
Determine the highlighted item
Determine the setting value
Returning a setting to its
original state before being
Returning to the previous
Switching the image mode
between the through mode
and freeze mode
Start measurement input
Switching the brightness level
(between H and L)
4 Image change key (SEL)
5 Measurement start/brightness select key
2 Set (select) key (SET)
3 Cancel key (ESC)
Select an item with the up, down, left and right
Set each coordinate.
- Select a digit or an item with the left and right
keys, and then specify a value with the up and
- Specify a value with the up, down, right and left
Switch the image mode between through and
freeze on the setting screen. (See the
preceding page.) When the mode is switched
from through to freeze, a new image is
Note: The "Gain/offset adjustment" screen is
only displayed in the through mode.
Press this key on the MAIN OPS MENU, and a
new measurement is triggered.
Change the displayed image brightness.
- Use this key when the image is so bright that
characters are difficult to see.
- The brightness can be changed on any
screen other than the run screen.