# The Individual Numerical Calculations For Each Measuring Program; The Setting Details - Sharp IV-S30 User Manual

Compact image sensor camera function and operation

Numerical Calculations

## 15-1 The individual numerical calculations for each measuring program

###  The setting details

The setting details and procedures for numerical calculations are the same for all measurement
programs with the exception of the point measurement program. In this section we explain the use of
the numerical calculations menu [NUMERIC CALC] for positional measurements.
- The [NUMERIC CALC] (numerical calculations) menu for positional deviation measurement.
[NUMERIC CALC]
1CALC RESULT N00(0~15) NO YES
2TYPE
3FORMULA
4FUNCTION
5UPR&LOW LIMIT +0000000.0000~+0000000.0000
6OUTPUT
7RUN A TEST
N00
[C]
N01
[X]
*1
N02
N03
1CALC RESULT:
(calculation results)
2TYPE:
3FORMULA:
(registation number)
Item2 selections
COORD [X Y], DEVIAT [x y], MATCH [M]
ANGL [B]
NUM-CAL [NC]
15
Constant C
Note: Please enter a smaller number for N (00 to 15) in the formula than the N (00 to 15)
in item1CALC RESULT
4FUNCTION
5UPR&LOW LIMIT:
6OUTPUT:
7RUN A TEST:
(make a test run)
COORD[X Y] DEVIAT[x y] MATCH[M]
ANGL[B] NUM-CAL[NC] CNST[C]
REG0(0~7) MDL0(0~1) N00(0~15)
CNST0000.0 + -
/
DEL END
*
SUM AVG REG0~0(0~7)
NO Y0(0~7) C000(0~107)
(SET KEY)
*3
*2
+0000002.0000
0X0
N 00
*
+0000400.0000~+0000410.0000 +0000406.0000
*4
You may set the location where the arithmetical operation results of the specified
output data will be stored, from locations 00 to 15.
-The results will be displayed 4 as one unit at position * 1 above.
Select the type of data being calculated.
A number of style settings are used.
- The style type is selected using the left and right keys and the numerical values
are entered using the up and down keys.
-The settings are displayed at N00 to N15, position * 3 above.
- The first line of the display will change, according to the selection made in item 2.
When obtaining the total (SUM) or the average (AVG) of the calculated results for
each object type 2, specify the range of register numbers.
Enter the upper and lower limits for making a judgment. -This setting will be
displayed in N00 to N15, at position * 4 above.
Setting the output of the calculation results.
The output can be set to Y0 to Y7 or C000 to C107. -This setting is displayed in
N00 to N15, at position * 6 above.
Pressing the SET key will store the setting details as well as run a test. -The
numerical results of the tests will be displayed at position *5 and the OK or NG
judgment will be displayed at position * 7.
Y0
OK
*5
Selection on line 3
REG0 (0 to 7) MDL0 (0 to 1) N00 (0 to 15)
CNST0000.0 + -
/
MEAS0 (0 to 7), N00 (0 to 15)
CNST0000.0 + -
/
ABS
TAN ATAN N00(0 to 14)
MAX MIN N00 to 00 (0 to14)
+0000000.00 (second line is not displayed)
15-2
The display will be different for
each measurement program.
-Page 15-6 to 9
Ex.:
The coordinates [X] is multiplied
by 2 and the results is stored in
N01
1. The constant 2 is stored in
N00
2. The X1 value of the registra-
tion number and the value in
6*
7*
Varies with each
DEL END
measurement
program.
-Page 15-6 to 9
DEL END
These are the
same as for the
other
measurement
programs.