1. Cut connector path
3. Turn board over
Installing the Instrument
Cut the existing conductor path on the upper side of the connector board with a
Very carefully remove all copper or plastic shavings from the board.
Turn board over.
Build the new connection to the Open-On-Alarm contact, shown below.
Using a piece of wire and solder, connect the three (3) points on the bottom of the
circuit board as illustrated in the following photographs:
Modification for Nurse Call Alarm Relays
2. Remove Shavings
4. Build new connection