Preparing to Install the Monitor
Explanation of symbols used:
Standby for switching the Monitor on and off.
Attention, consult accompanying documents.
Infra-red Connector for connection to a printer.
On the Measurement Server - Defib Data In, that is the ECG marker
pulse sent from the defibrillator to the Monitor. The marker pulse is then
processed with the ECG signal and displayed on the Monitor.
On the M3015A Measurement Server Extension - Gas Input
On the Measurement Server: ECG Data Out is the analog ECG signal
sent out from the Monitor to a defibrillator or other external device, such
as an intra-aortic balloon pump.
On the M3015A Measurement Server Extension: Gas Outlet/Exhaust.
On the Monitor: Data Output to serial recorder
Serial interface socket for connection to recorder (M3046A #J16 only)
Equipotential Grounding Post (see Equipotential Grounding earlier in this
Type CF Applied Part and defibrillator proof with special protection
against electric shocks for intracardiac application (regarding allowable
leakage currents by having an F-Type isolated or floating section).
Indicates a Monitor with Wireless LAN Interface (symbol appears on
Class 2 Radio equipment identifier (1999/5/EC)
Installing the Instrument