Refit Procedures for the M3015A Measurement Server
• A thin-bladed screwdriver.
• A pair of large tweezers.
• In addition, for refitting the Pump, you will need a large-bladed screwdriver.
There is high voltage inside the Instrument (800V). Do not connect the Measurement
Server Extension to a Monitor while the Extension housing is open.
As well, parts inside the Instrument may be contaminated with bacteria. Protect
yourself from possible infection by wearing examination gloves during these
Refitting the CO
not open or damage the CO
material, flush the area immediately with water and consult a doctor.
To refit the CO
Repairing the Instrument
Refit Procedures for the M3015A Measurement Server Extension
Scrubber contains lithium hydroxide monohydrate. This is a strong base. Do
Scrubber. If you come into contact with the CO
Scrubber, do the following:
Feed the end of the CO
Scrubber through the bracket to meet the Extension intake
Push the intake tube firmly into the scrubber end to connect it.
Holding the body of the CO
air intake under the second bracket and position it.
Scrubber with tweezers, feed the CO