The mounting solutions described in this manual are subject to change. For the latest
information, please visit the Patient Monitoring section of the Philips Medical Systems
website, and follow the links to mounting solutions.
For detailed mounting instructions, refer to the documents supplied with the mounting
Installing the Table Mount (M3080A #A10)
The Table Mount is intended for any flat, level surface. To mount the plate securely, you need
4 screws. These are not provided because the type of surface dictates the type of screw.
The Monitor snap-fits onto the plate. To release the Monitor, press the button on the front of
This mount is also:
• available as support part M3080-63201
• used for the Rollstand mount M3080A #A22
Unpack the Table Mount and check all parts are available.
Select 4 screws (appropriate to surface material).
Insert the 4 screws through the screw holes provided in the plate.
Tighten the screws.
Installing the Instrument