Installation of Wireless Infrastructure
If this file does not automatically appear in the Configuration File field, click Browse and
find this file on the computer drive directory where it was stored.
c. click Browse to access the Browse application
d. select ConfigTool in the tools menu of the stored configuration tools files
e. locate the file ConfigTool.exe in the ConfigTool directory
f. double click ConfigTool.exe to enter it into the Open: field of the Run
g. click OK to open the Configuration Tool window:
The Configuration Tool Window is used to select the Serial Port (COM1 or
COM2) on the configuring PC that will be used to perform the configuration and
the configuration file for the device being configured -- Access Point, Network
Switch, or Wireless Bedside.
Select the Serial Port to be used (COM1 or COM2) in the Select Serial Port field
by clicking in the circle preceding the appropriate port.
Select the appropriate Device to Configure in the drop down list in the
Configuration Tool window of the configuring PC. See previous figure.
Select the file tools\ConfigTool\WB_WirelessM3_1-2b15.cfg in the
Configuration File fields.
Turn OFF the power of the Wireless M3/M4 Monitor and disconnect any cable
connected to the RJ-45 port on its rear panel.
Installing the Instrument