Cleaning and Maintenance
Performing cleaning and maintenance regularly reduces the chances of equipment failure.
CAUTION! Personnel performing any maintenance or cleaning must familiarize
themselves with the information in
before attempting any of these procedures.
Disconnect cable (if connected) and turn device off by pressing and holding the
for approx. 3 s.
Clean all surfaces using a soft, lint-free cloth with a mild soap solution. Avoid getting
moisture in openings.
Replacing filter, needle and measuring gas hose
During use of the device, dust will over time be sucked through the measuring gas system,
and therefore the filter
fluid from the packaging from which the measurement is taken.
If this happens, it will be necessary to replace both the filter
If there is any dirt or fluid in the measuring gas hose
dry compressed air.
CAUTION! When using compressed air for cleaning, the hose must be disconnected
from the device.
will have to be replaced regularly.
may sometimes touch the product being measured or the needle may suck
"Safety and handling instructions" on page 75
, the needle
, it should be replaced or cleaned using
, and the hose