This technical manual provides operating instructions for the Air Data Test System compatible
with the requirements of first line operation for the technician and supervisor.
This technical manual contains a brief description, operation and testing procedures for the
user of this equipment with software version V1.08.
The manufacturer has designed this equipment to be safe when operated using the
procedures detailed in this manual. Do not use this equipment for any other purpose than that
stated, the protection provided by the equipment may be impaired.
This publication contains operating and safety instructions that must be followed to make sure
of safe operation and to maintain the equipment in a safe condition. The safety instructions are
either warnings or cautions issued to protect the user and the equipment from injury or
Use qualified* technicians and good engineering practice for all procedures in this publication.
The ADTS 505 contains two protected menus, the operating limits (described in this manual)
and the maintain calibration menu (described in the calibration manual).
The factory set PIN codes are contained in an envelope addressed to the Supervisor.
Change these codes for authorised access. Unauthorised access to these two menus can make
this system inaccurate and could, in control mode, cause excessive rates of pressure change.
Do not apply pressure greater than the maximum safe working pressure to the equipment.
There are no known toxic materials used in this equipment.
The equipment must be maintained using the manufacturer's procedures and should be
carried out by authorized service agents or the manufacturer's service departments
For technical advice contact the manufacturer or subsidiary.
A qualified technician must have the necessary technical knowledge, documentation, special test
equipment and tools to carry out the required work on this equipment.
K0260 Issue No. 8