Druck ADTS 505 User Manual
Selected in Leak/Measure Mode
Use the touch screen keys to select wait and measure times and use the numeric keys to enter the
Note: Times entered as whole numbers are interpreted as seconds. Numbers higher than 60 are interpreted
and displayed as minutes and seconds (i.e. 90 = 1 minute 30 seconds = 1:30).
After selecting the timer values and with the required the Ps1, Ps2 and Pt readings, press the START
key. The time counter shows the progress of the test: wait time, then test time. The results, at the end
of the test, shows the timed rates for each channel indicated by "T" after the digits.
Press BACK to return to Leak Measure Mode.
Alt and CAS
K0260 Issue No. 8
6 - 25