Druck ADTS 505 User Manual
Fuse Replacement (Fig 4-1)
Replace the fuse when detailed in Section 5, Testing and Fault Finding:
Set the power switch to OFF.
Note: The power switch is not a power isolator as defined by EN61010.
Isolate the power supply to the instrument by disconnecting the power supply
Unscrew and remove the fuse cap (2) and fuse cartridge (3) from the front panel and
replace the fuse cartridge (3).
Note: Fit the correct type of fuse detailed in Section 2, Installation and on the front panel.
Refit the fuse cap (2) and fuse cartridge (3) to the front panel.
Reconnect power supply connector (1).
Switch on the power supply. The instrument should now be operational.
If a fuse blows immediately on switch-on, contact the manufacturer or Service Agent.
K0260 Issue No. 8
4 - 5