2 - 2
The unit should be at zero/ambient pressure. Disconnect the hose assemblies and stow in the
Switch OFF and disconnect from the electrical power supply. Disconnect the power
supply cable and the hand terminal cable. Disconnect the hand terminal cable from the
Fit the lid to the unit.
The power supply cable, hand terminal cable and the hand terminal should be placed
in the original packing material.
If available, use the original packing material. When using packing materials other than the
original, proceed as follows.
Wrap unit in polyethylene sheeting.
Select a double-wall cardboard container. Inside dimensions must be at least 15 cm
greater than the equipment. The carton must meet test strength requirements of >125 kg.
Protect all sides with shock-absorbing material to prevent equipment movement within
Seal carton with approved sealing tape.
Mark carton "FRAGILE" on all sides, top, and bottom of shipping container.
To return the unit for calibration or repair complete the return goods procedure as detailed in 2.3.
The following conditions apply for both shipping and storage:
Temperature Range ........... ............-20 to +70°C (-4 to +158°F)
Altitude......... ............Up to 15,000 feet (4,570 metres)
K0260 Issue No. 8