Druck ADTS 505 User Manual
The ADTS 505 provides a power supply to the advanced hand terminal when connected to the
hand terminal connector on the front panel.
The power supplies must be isolated when connecting the advanced hand terminal in the
Using the advanced hand terminal when not connected to the ADTS 505
Using the advanced hand terminal with a pc to create test scripts.
The unit must be connected to the correct electrical power supply, see paragraph 2.4.
Before use, make sure the SELV power adaptor supplied with the instrument is correct for the
power supply voltage. The Safety Extra Low Voltage (SELV) power adaptor complies with
EN61010 (including safety requirements for laboratory instruments).
Note: The instrument can be powered from other DC power supplies of the correct voltage range. It
is the user's responsibility to make sure the power supply is safe.
Make sure that the power supply is off before connecting the power cable.
Communications cable - parts list item 11
Pneumatic Pressure Connections
Ps (static)- AN4 (MS33656-4)
When not in use, a blanking cap must be fitted.
Note: When carrying out a leak test, a leak of this blanking cap affects the performance of the ADTS 505.
Jack plug outer
Jack plug inner
2 - 7
K0260 Issue No. 8