Air outlet section
The air outlet section contains the following
• Compressed air reservoir
• Pressure manometer
• Moisture relief valve
• Compressed air outlet
• Compressed air inlet
• Cooling air outlet.
The dried air is supplied via water separator 1
in the dryer section to the compressed air
reservoir. The compressed air is available at
the compressed air outlet.
The air pressure in the compressed air
reservoir is displayed by the pressure
If water, for some reason, is collected in the
compressed air reservoir, this water can be
removed by pressing the valve pin of the
moisture relief valve.
Compressors equipped with the standby
function include a compressed air inlet
where the hospital central gas supply
(medical grade air) is connected.
The cooling air stream that has been forced
through the cooling section and compressor
section exits via the silenced cooling air
E382 EH81E 061 01 03 02