7.3 Handpiece / Cable
7.4 Cavitron 30K Ultrasonic Inserts
The many styles of Cavitron and Cavitron Bellissima 30K Ultrasonic Inserts are easily interchangeable for various procedures
and applications. See enclosed literature for specific information.
Provides seal for handpiece coolant.
O-ring should be replaced when worn.
Shape and size of tip
determines access and
Lavage directed to tip.
(YDDD=Single Digit Year
and Triple Digit Day of
the Year), Frequency,
Type, Tip Lot Number
Turn the Lavage Control to select flow rate during system
operation. Clockwise increases flow at insert tip, counter-
clockwise decreases flow. The flow rate through the
handpiece also determines the temperature of the lavage.
Lower flow rates produce warmer lavage. Higher flow rates
produce cooler lavage.
If the handpiece becomes warm, increase the flow rate.
With experience the Dental Healthcare Professional will be
able to determine the best flow rate setting for optimum
operating efficiency and patient comfort.
Reduces cable drag as handpiece rotates during procedures.
Soft Nozzle Grip
Provides an ergonomic and comfortable grasp of the
handpiece. The grip is a replaceable wear component. Prior
to use, verify that the grip is flush with the hard plastic of the
Powder Delivery Port
Creates an airtight seal between the air polishing insert and
the handpiece. Replace when wear is noticed or powder is
leaking at nozzle interface.
The Cavitron® JET-Mate Sterilizable Handpiece accepts all
30K Ultrasonic Inserts and JET air polishing insert.
Transfers and amplifies mechanical
motion of stack to insert tip.
Converts energy provided by the handpiece
into mechanical oscillations used to activate
the insert tip.