Ultrasonic Scaling System Détartreur ultrasonique Escarificador ultrasónico Ultraschall-Zahnsteinentfernungsgerät Scaler ultrasonico Ультразвуковая система для удаления зубного камня Directions For Use Mode d’emploi Instrucciones de uso Gebrauchsanweisung Istruzioni per l’uso Указания по применению Please read carefully and completely before operating unit. Prière de lire attentivement et complètement avant la première utilisation de l’appareil. Por favor lea cuidadosamente y en su totalidad antes de operar la unidad.
– the health of each patient, – the dental procedures being undertaken, – and applicable industry and governmental agency The Cavitron Touch™ Ultrasonic Scaling System offers an recommendations for infection control in dental advanced experience at your fingertips. The unit features a...
Steri-Mate ® handpiece are clean the Cavitron Touch Ultrasonic Scaler will create aerosols. and dry before assembling them for use. Following the procedural guidelines in Section 9 of this manual can effectively control and minimize aerosol 4 2 procedural precautions dispersion.
• A water supply line with user-replaceable filter is supplied with your system. See Section 10 System Care • The Cavitron Touch System is designed to rest on a level for replacement instructions. surface. Be sure unit is stable and resting on four feet.
stopped rotating and sound indication occurs. 7 6 water Supply Line Connection 9. To verify proper communication, press the Scale icon to return to the Main screen (scale screen). Depress the foot • Grasp the Water Supply Line (blue hose) by the end pedal to the second position and ensure that Boost is opposite the quick-disconnect and insert it into the water activated (seen in Power Level icon).
SECTiON 8: Cavitron Touch ™ ultrasonic Scaler description 8 1 System Controls Ultrasonic Power Level Control Slide to select the desired ultrasonic power level for operation. Sliding up increases the power level and sliding down decreases the power level. Rinse Rinse mode is for use during an ultrasonic scaling procedure when lavage is desired with no cavitation (Page 8).
8 2 diagnostic display indicators and Control Service Foot Pedal Battery 1 - Cable or handpiece issue. Indicates battery life of foot pedal. 2 - Unit overheating. Stop use and contact Cavitron Care. Preset Power Modes- Icon will appear when water filter Programmable needs to be replaced.
8 2 diagnostic display indicators and Control, continued Scale Press to return to main screen Settings Screen Tap-On Press to activate or inactivate Tap-On™ Technology Note: Tap-On is inactivated at the factory. Sync Press to synchronize foot pedal and unit Note: If the foot pedal is synced to the unit, pressing the sync icon will unlink the foot pedal from the unit.
8 3 handpiece / Cable Lavage Control Turn the Lavage Control to select flow rate during system operation. Flow rate is based on a scale from 1 to 6. Turn clockwise toward 6 to increase flow at insert tip. Turn counter-clockwise toward 1 to decrease flow.
Boost is still available while scaling in Tap-On mode. To use Boost, simply depress the For detailed information, contact your local DENTSPLY foot pedal to the second position (all the way to the floor) to Representative or authorized DENTSPLY Distributor.
cases, are adequate to remove even the most tenacious 9 2 Boost mode calculus. Boost provides a temporary increase in ultrasonic scaling • Periodically check the Cavitron Ultrasonic Insert for wear power for quick removal of tenacious calculus without with the Cavitron Insert Wear Indicator. touching the unit.
6. Inspect the handpiece cable for any breaks or tears. reduced water flow to the Cavitron Touch system. If using a closed water supply or DualSelect Dispensing 1. Verify that the system is turned OFF.
SECTiON 11: Troubleshooting Symptom: System operates: No insert cavitation Although service and repair of the Cavitron Touch Ultrasonic Check the insert for damage and that it is properly Scaler should be performed by DENTSPLY personnel, the installed in the handpiece.
SECTiON 12: Atmospheric Pressure: 500 to 1060 hPa warranty period The Cavitron Touch Ultrasonic Scaler is warranted for TWO YEARS from date of purchase. The Steri-Mate 360 Handpiece enclosed with your system is warranted for SIX MONTHS from date of purchase. Refer to the Warranty Statement Sheet furnished with your system for full Warranty Statement and Terms.
SECTiON 14: Symbol identification MEDICAL EQUIPMENT AC POWER WITH RESPECT TO ELECTRIC SHOCK, FIRE AND MECHANICAL HAZARDS ONLY IN ACCORDANCE WITH UL 60601-1CAN/CSA-C22.2 No. 601.1, TYPE B APPLIED PART EQUIPMENT ANSI/AAMI ES60601-1 (2005, 3rd ed.) CAN/CSA-C22.2 No. 60601-1 (2008), 13VA PROTECTIVE EARTH (GROUND) This device complies with part 15 of the FCC Rules.