Read This First
System users are responsible for image quality and diagnosis. Inspect the data that is being
used for the analysis and diagnosis and ensure that the data is sufficient both spatially and
temporally for the measurement approach being used.
Before using the system, read these warnings and the
Do not remove the protective covers on the system; hazardous voltages are present inside.
Cabinet panels must be in place while the system is in use. All internal adjustments and
replacements must be made by a qualified Philips Ultrasound field service engineer.
To avoid electrical shock, use only supplied power cords and connect only to properly
grounded wall (wall/mains) outlets.
Do not operate this system in the presence of flammable gases or anesthetics. Explosion can
result. The system is not compliant in AP/APG environments as defined by IEC 60601-1.
Medical equipment must be installed and put into service according to the special
electromagnetic compatibility (EMC) guidelines provided in the
The use of portable and mobile radio-frequency (RF) communications equipment can affect
the operation of medical equipment.
EPIQ 7 User Manual 4535 617 25341