Manipulating the S7-3t TEE Tip
Review the warnings and caution in
Deflection Control Basics " on page 207
1. Put the deflection control knob into freewheeling mode by sliding the deflection lock so
that the indicator is white.
2. Do one of the following:
3. Once the tip is positioned properly, lock the deflection control knob by sliding the deflection
lock so that the indicator is red.
Manipulating the S7-3t Transducer Tip
To deflect the tip toward the posterior, turn the smaller knob clockwise.
To deflect the tip toward the anterior, turn the smaller knob counterclockwise.
To set the tip in the neutral position (no deflection), turn the smaller knob so that its
white bar is in line with the center of the proximal end of the transducer handle.
"Patient Safety During TEE Studies" on page 200
before using the transducer in a study.
Using the S7-3t TEE Transducer
EPIQ 7 User Manual 4535 617 25341