Using the X7-2t TEE Transducer
Manipulating the X7-2t Tip
Review the warnings and caution in
Deflection Control Basics " on page 207
1. Turn the detent brake actuator fully away from the image plane rotation buttons to put
both knobs into freewheeling mode.
2. Turn the large knob to deflect the tip in the anterior/posterior plane.
3. Turn the small knob to deflect the tip in the medial/lateral plane.
4. When the tip is positioned properly, do one of the following:
EPIQ 7 User Manual 4535 617 25341
Turn the detent brake actuator fully toward the image plane rotation buttons to put
both knobs in detent mode.
Center the detent brake actuator to put only the small knob (medial/lateral
movement) in the detent mode.
Image plane rotation buttons
Neutral position indicators, showing no deflection
"Patient Safety During TEE Studies" on page 200
before using the transducer in a study.