MAN0443.P65 Issue 13 Aug 04
CHAPTER 8 - SPECIFICATION
APPROVALS AND STANDARDS
Designed to comply with:
General Requirements (Combustible Gases).
Performance (100 LEL).
Performance (100 V/V).
Software and Digital Technologies.
Meets Exe isolation requirements for 50V operation.
EXAM BBG Pruef- und Zertifizier GmbH, EC-type examination certificate BVS
04 ATEX G 001 X.
Note: If compliance with the EC-type examination certificate BVS 04
ATEX G 001 X is required, refer to chapter 10 'Special Conditions
for Safe Use'. Ensure that all conditions therein are met.
EMC Directive 89/336/EEC
The optional recorder output may be in error by ±5% fsd during some
test conditions of the industrial standard if the connecting cables are
not shielded with an outer screened cable.
Note: The Part 2 conformity refers to installations using the System 57
cabinet. For System 57 racks not supplied in cabinets or supplied
in GRP cabinets, the conformity is to Part 1.
C* to +55
(*0° for EXAM approved systems).
C to +55
0 to 90% RH. Non-condensing.
10V/m over 50kHz to 1GHz.
8 - 3
5701 Control System