For transmitters rated at a maximum output power not listed above, the recommended separation dis-
tance d in metres (m) can be estimated using the equation applicable to the frequency of the transmitter,
where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by
absorption and reflection from structures, objects and people.
• Guidance and manufacturer's declaration – electromagnetic immunity
The device is intended for use in the electromagnetic environments listed below, and should only be
used in such environments:
Electromagnetic environment – guidance
Portable and mobile RF communications equipment should be
150 kHz to
used no closer to any part of the device, including cables, than
the recommended separation distance calculated from the
equation applicable to the frequency of the transmitter.
Recommended separation distance
d = 1.2
80 MHz to
d = 2 .3
where P is the maximum output power rating of the transmitter
in watts (W) according to the transmitter manufacturer and d is
the recommended separation distance in metres (m).
Field strengths from fixed RF transmitters, as determined by an
electromagnetic site survey, a should be less than the compli-
ance level in each frequency range.b
Interference may occur in the vicinity of equipment marked
with the following symbol:
NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorp-
tion and reflection from structures, objects and people.
a Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and
land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted
theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an
electromagnetic site survey should be considered. If the measured field strength in the location in which the
device is used exceeds the applicable RF compliance level above, the device should be observed to verify
normal operation. If abnormal performance is observed, additional measures may be necessary, such as
reorienting or relocating the device.
b Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 10 V/m.
80 MHz to 800 MHz
800 MHz to 2.5 GHz
If any abnormality arises during use, please check the following points.
Unit does not turn
on when the START/
STOP button is
have been positioned
EE measurement error
Cuff has been placed
symbol shown on
display or the blood
pressure value is
Did you talk or move
low (or high).
Shaking of the arm with
the cuff on.
E1 error symbol shown
Air circuit abnormality.
Cuff tube may not be
plugged into monitor
E2 error symbol shown
exceeding 300 mmHg.
E3 error symbol shown
EP error symbol shown
Note: If the unit still does not work, contact HoMedics Consumer Relations.
Under no circumstance should you disassemble or attempt to repair
the unit by yourself. Contact information for HoMedics Consumer
Relations Department can be found on the warranty page.
Replace them with
four new "AAA"
Re-insert the batteries in
the correct positions.
Wrap the cuff properly so
that it is positioned
correctly. Measure again.
Keep arm steady during
Check cuff connection.
Switch the unit off, then