To avoid electric shock or injury, do not perform the performance tests or
calibration adjustment procedures unless qualified to do so.
The information provided in this document is for the use of qualified personnel
The 114, 115, 116, & 117 Calibration Information provides the information necessary to adjust and verify the
performance of the Fluke Models 114, 115, 115C, 116, 116C, 117, and 117C True RMS Multimeters (hereafter
known as the Meter). When specific models are noted in this manual, the "C" version is also included. For example,
when the listed model is 115, the instructions are applicable to the 115C as well.
The following information is included in this document:
Safety Information and International Electrical Symbols (page 2)
Specifications (page 3)
Testing the Fuse (page 5)
Replacing the Fuse and the Battery (page 6)
Cleaning (page 7)
Performance Tests (page 7)
Calibration Adjustment (page 12)
Replacement Parts and Accessories (page 17)
Complete Warranty (page 19)
See the 114, 115, 117 Users Manual or the 115C, 117C Users Manual or the 116 Users Manuals or the 116C Users
Manual for complete operating instructions.
To contact Fluke, call one of the following telephone numbers:
USA: 1-888-99-FLUKE (1-888-993-5853)
Canada: 1-800-36-FLUKE (1-800-363-5853)
Europe: +31 402-675-200
Anywhere in the world: +1-425-446-5500
Or, visit Fluke's Web site at www.fluke.com.
To register your product, visit
September 2006 Rev. 1, 10/08
©2006, 2008 Fluke Corporation. All rights reserved. Specifications are subject to change without notice.
114, 115, 116, and 117