Mini-AT User Guide
Volume Function Check
While in the Corrector Mode, note the corrected volume reading on the LCD. Rotate the index input wriggler
in one direction for ten (10) revolutions. Verify that the volume has increased by an expected amount. With zero
applied pressure and at room temperature, the additional corrected volume should be essentially equal to the
meter index rate, times the number of turns (10) of the wriggler. Note: the scaling of corrected volume (item 090)
may make it necessary to apply more than ten revolution to see an increase at the LCD.