Indications for Use/Intended Use Symbols on the product The LCD Monitor is intended to provide 4K, 3D and 2D Safety sign (LMD-X310MT only) color video displays of images from endoscopic/ Follow the warnings in the instructions for use laparoscopic camera systems, surgical microscope and for parts of the unit on which this symbol other compatible medical imaging systems.
All interface cables used to connect peripherals must be different branch circuits. shielded in order to comply with the limits for a digital For more information, consult qualified Sony service device pursuant to Subpart B of part 15 of FCC Rules. personnel.
• Portable and mobile RF communications equipment, such as cellular phones, can affect the LMD-X310MT/X550MT. WARNING The use of accessories and cables other than those specified, with the exception of replacement parts sold by Sony Corporation, may result in increased emissions or decreased immunity of the LMD-X310MT/X550MT.
Guidance and manufacturer’s declaration – electromagnetic immunity The LMD-X310MT/X550MT is intended for use in the electromagnetic environment specified below. The customer or the user of the LMD-X310MT/X550MT should assure that it is used in such an environment. Compliance level IEC 60601 Electromagnetic environment –...
Guidance and manufacturer’s declaration – electromagnetic immunity The LMD-X310MT/X550MT is intended for use in the electromagnetic environment specified below. The customer or the user of the LMD-X310MT/X550MT should assure that it is used in such an environment. IEC 60601 Compliance Immunity test Electromagnetic environment –...
Recommended separation distances between portable and mobile RF communications equipment and the LMD-X310MT/X550MT The LMD-X310MT/X550MT is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of the LMD-X310MT/X550MT can help prevent...
LMD-X550MT only WARNING Warning on power connections for To avoid the risk of electric shock, this medical use equipment must only be connected to a supply Customers in the U.S.A. and Canada should use the mains with protective earth. following type of power cord. Customers in other countries or regions should use the WARNING power cord prescribed by their country or region.
Disposal of Old Electrical & Electronic Equipment (Applicable in Republic of India) For the customers in the U.S.A. SONY LIMITED WARRANTY - Please visit http:// www.sony.com/psa/warranty for important information and complete terms and conditions of Sony’s limited warranty applicable to this product.
(a) Before actually using this device for medical LMD-X310MT is a DC powered device. Use with the supplied AC adaptor (AC-300MD). practice, please check and confirm that you do not LMD-X550MT is an AC powered device.
“stuck” pixels may appear spontaneously. These Masked images with aspect ratios other than 17:9 for problems are not a malfunction. the LMD-X310MT and 16:9 for the LMD-X550MT Do not leave the LCD screen facing the sun as it can ...
LMD-X310MT use of CFV-E30SK. LMD-X550MT The illustration of LMD-X310MT is used for the Before cleaning explanations. Any differences in specifications are Be sure to disconnect the AC power cord from the AC clearly indicated in the text. outlet.
Compliance with medical safety standards A.I.M.E. is a registered trademark of Sony Corporation. A.I.M.E. is an optional function for enhancing the color in U.S.A., Canada and Europe or structure of the displayed image.
Flat surface for better maintenance The design allows the user to easily wipe liquids and gel off the LCD panel and control buttons – ensuring a high standard of disinfection and cleanliness. Grip that is easy to hold The easy-to-grip ergonomic design enables the user to make tiny adjustments simply and quickly.
Location and Function of Parts and Controls Front Panel Power indicator Turns on or off the assigned function. You can adjust the assigned function by pressing the / buttons. (Refer to When the power is turned on, the power indicator lights the custom buttons of the system configuration menu on in green.
displayed, a menu which allows you to select the input (On)/ (Standby) switch signal to be assigned to PORT A is displayed. Press the side to turn the monitor on. Press the side PORT B: Displays the input signal assigned to to switch the unit to the standby state, and the indicator PORT B.
SD-SDI: 200 m (approx. 656 ft) max. (When using 5C-2V Input connector for HDMI signals. coaxial cables (Fujikura. Inc.) or equivalent.) Use HighSpeed HDMI Cable with Cable Type Logo. (Recommend Sony Cable.) 3G/HD/SD-SDI input/output connector (BNC type) SDI 1 (4K/HD)
Signals input to the input connector are output to the output connector. (DC 26 V input) connector (LMD-X310MT only) 1) Serial digital component signals are SMPTE ST 259 Connects the DC connector of the supplied AC adaptor.
The connectors on this unit are not isolated. Do not connect any device other than one which conforms to IEC 60601-1 standards. LMD-X310MT When an information technology device or AV device that uses an alternating current is connected, current Make sure that the (On)/...
Connect the plug of the AC power cord to an AC outlet. To remove the AC power cord Press the (On)/ (Standby) switch to the side and switch the unit to the standby state, then remove the AC Connect the plug of the AC power cord to an AC power cord from the AC power plug holder by holding outlet.
Caution This unit with the connector cover attached conforms to Remove the two screws and cable clamp. the waterproofed standard. (LMD-X310MT: See page 34, LMD-X550MT: See page 35) Be sure not to use the unit with the connector cover removed since waterproofed performance is not Caution guaranteed.
Configuration Example: Initial Setting 4K Surgical Endoscope When you turn on the unit for the first time after purchasing it, select the area and language where you The 4K monitors can be configured with a 4K endoscope intend to use this unit from among the options. and 4K recorder as below.
Press the or button to select the area where you intend to use the unit and press the button. When the confirmation screen is displayed, press the or button to select Yes and press the CONTROL button.
Using the Menu The unit is equipped with an on-screen menu for making various adjustments and settings such as picture control, input setting, setting change, etc. To change the menu language, refer to “To change the menu language” on page 24. Press the /...
The selected tab is shown in blue, and setting items Adjustment Using the of the selected tab appear. Menus Select an item. Press the / button to select the item. The item to be changed is displayed in blue. Items Make the setting or adjustment on an item.
Pattern Skip Submenu Setting Contrast Adjusts contrast. Input Configuration A.I.M.E. Select from among “Off, ” “On” and “Check Mode. ” Input Name When you select “Check Mode, ” two HDCP Setting images of A.I.M.E. being “Off” / “On” are displayed. 1) A.I.M.E.: Enables better image System Configuration reproduction and display.
Screen Control menu Submenu Setting The Screen Control menu is used to set the image display Interface Mode Select the interface mode. setting for each input. 4K Quad: Select to display the 4K signal input to the A to D connector of SDI 1.
Submenu Setting Submenu Setting 3D Signal Format Select the 3D signal format. Port C Input Sets the input connector for the third Dual Stream: Select to display signals screen. of the Dual Stream system in 3D. Select from among “SDI 1, ” “SDI 2, ” Side by Side: Select to display signals “DVI-D, ”...
Input Configuration menu Submenu Setting Submenu Setting Select Port Input connector Control lock display Input Name Sets the name for each input connector. Set the name of the SDI 1, SDI 2, DVI- D, HDMI connector. Signal Format display Endoscope ...
Submenu Setting Submenu Setting Language You can select the menu or message Custom Button Assigns the function to the CUSTOM 1 language from the following languages. or CUSTOM 2 button on the front English: English panel, and can set the following ...
Submenu Setting Troubleshooting PIP / POP PIP Sub Screen Sets the sub screen position for the 2 or Position 3 screen displays (PIP). This section may help you isolate the cause of a problem Select from among “Lower Left, ” “Upper Left, ”...
Error Messages Specifications When the following messages appear on the screen, turn LMD-X310MT off the power and contact an authorized Sony dealer. Messages Description Picture performance Fan Error A malfunction occurs in the fan. LCD panel a-Si TFT Active Matrix...
Operating conditions Medical Specifications Temperature Protection against harmful ingress of water: 0 °C to 40 °C (32 °F to 104 °F) IPX2 Recommended temperature Degree of safety in the presence of a flammable 20 °C to 30 °C (68 °F to 86 °F) anesthetic mixture with air or with oxygen or Humidity 30% to 85% (no condensation)
3D Eye Shield Kit (CFV-E30SK) (1) Always verify that the unit is operating properly Frame (1) before use. SONY WILL NOT BE LIABLE FOR 3D shield (3) DAMAGES OF ANY KIND INCLUDING, BUT AC power plug holder (2) NOT LIMITED TO, COMPENSATION OR...
Pin assignment SERIAL REMOTE (RS-232C) connector D-sub 9-pin, female Pin number Signal Specifications...
Available signal formats The unit is compatible with the signal systems shown below: Signal format SDI 1 SDI 2 SD-SDI 1) 7) 720 × 487/60I 4 : 2 : 2 YCbCr 10bit 720 × 576/50I 4 : 2 : 2 YCbCr 10bit HD-SDI ...
Dot clock: 25.175 MHz to 148.5 MHz Picture size, phase: automatic discrimination by the DE (Data Enable) signal Displays normally up to the maximum horizontal resolution, 2048 dots (LMD-X310MT) or 1920 dots (LMD- X550MT). 1) Also compatible with the frame rate 1/1.001.
LMD-X310MT/X550MT 3D Viewing Angle (vertical) * Length of M4/M6 screws Attached object Monitor cabinet Screw ) to 9 ( * The specification of the screw is for the VESA mount. Side View Unit: mm (inches) The supplied screws are compatible with an attached...