If any abnormality arises during use, please check the following points.
Unit does not turn on
Batteries have run
when the START/STOP
button is pushed.
Battery polarities (+/-)
have been positioned
EE measurement error
The UltraSoft™ upper
symbol shown on
arm cuff has been
display or the blood
positioned on the arm
pressure value is
Did you talk or move
low (or high).
Shaking of the arm with
the cuff on.
E1 error symbol shown
Air circuit abnormality.
Cuff tube may not be
plugged into monitor
E2 error symbol shown
exceeding 300 mmHg.
E3 error symbol shown
EP error symbol shown
Note: If the unit still does not work, contact HoMedics Consumer Relations.
Under no circumstance should you disassemble or attempt to repair the
unit by yourself. Contact information for HoMedics Consumer Relations
Department can be found on the warranty page.
Replace them with four new
"AA" alkaline batteries.
Re-insert the batteries in the
correct positions. See page 11.
Re-wrap the cuff properly so
that it is positioned correctly.
Take new measurement.
Keep arm steady during
again. Refer to "Measurement
Procedure" instructions, see
Check cuff connection.
See page 15. Take new
Switch the unit off and then
turn back on. Take new
Re-wrap the cuff properly so that
it is positioned correctly. Take
Remove batteries. After 1
minute, re-insert the batteries.
Take new measurement.
Four 1.5V DC "AA" alkaline batteries
or 6V DC AC adapter
Measurement Method: Oscillometric
Pressure: 40~280 millimeters mercury (mmHg)
Pulse: 40~199 beats/minute
Pressure: ±3 mmHg
Pulse: ±5% of reading
Automatic Pressure release valve
60 memories for each User (120 total)
1 minute after last button operation
Temperature: 41°F~104°F (5°C ~40°C)
Humidity: 15% ~ 93% RH
Temperature: -13°F~158°F (-25°C~70°C)
Humidity: Less Than 93% RH
0.69 lbs (310 g) (without batteries)
Upper Arm Cuff size: 9"–17" (23-43 cm)
4.21"(L) x 5.55"(W) x 2.48"(H)
107mm(L) x 141mm(W) x 63mm(H)
6V DC AC adapter, 1 arm cuff with tube
Internally Powered equipment
Type BF Equipment
IPx0-No special protection against the
ingress of water
Not suitable for use in presence of flammable
anesthetic mixture with air, oxygen or
Continuous operation with short-time loading.
Follow instructions for use.
Note: These specifications are subject to change without notice.
11/18/14 12:41 PM