Install a switch and fuse to isolate the
LF414/LF600F, LF610F and LF414 from
Power supply from mains power can
cause electric shock or circuit
Do not work on piping and wiring with wet
Wet hands may result in electric
Do not conduct wiring work with bare hands.
Remaining electric charge even if
power is turned off can still cause
Do not modify or disassemble the LF600F,
LF610F, LF602F and LF612F
Modifying or disassembling this
product can cause electric shock,
malfunction of or damage to this
Flowmeter accuracy may be affected by the way wiring is executed. Proceed with correct wiring taking
the precautions in following pages.
Turn off mains power before conducting wiring
Wiring while power is applied can cause
Ground the LF600F, LF610 independently from
power equipment. (100 ohm or less ground
Operating this product without grounding
can cause electric shock or malfunction.
For the power supply wiring and grounding wiring,
use crimping terminals with insulated sleeve.
There is a risk of electric shock due to
drop-off or loosing, and a risk of fire and
equipment trouble due to heat generation.
The label shown left is placed near the
power supply terminal on the
Be alert to electric shock.
－ ３１ －