Cleaning and maintenance work
Routine maintenance tasks must be performed at the intervals shown in the table.
It is however advisable to carry out simple cleaning whenever necessary.
In case the machine requires the replacement of one or more components, please refer to the
manufacturer's spare part list.
Cleaning of casing
Cleaning of external surfaces
Spray nozzles: Check and clean as necessary.
Check disinfection performance.
Filters in valves: Check, clean and replace as
Check safety thermostats.
Chemical dispenser pumps: Check the membrane and
Level fill sensors for process chemicals: Check
function and clean hoses.
Dispenser pump hoses: Check for kinks, leakages or
Check function of level sensors in water tank.
Temperature probes: Periodically check sensor status
during maintenance work.
Drain valves: Check for leakage, if necessary remove
and clean the membrane seat, and replace if
Pressure switches: Operation is monitored by the
controls. In the event of defective controls, empty the
tank and remove deposits.
It is advisable to perform general checks and to clean the machine regularly,
particularly if the water supply is very hard.
Particular attention should be paid to heater elements and thermostats sensors.