ECG EQUIP MALF INOP
This INOP indicates that either a software incompatibility
has been found or a fault has been detected in the ECG
hardware in the transmitter.
1. If the transmitter is an EASI ä ECG transmitter, this
INOP indicates that the software in either the
receiver mainframe or the central station cannot
process EASI ä ECG data. Do one of the following:
a. Update the software in the receiver mainframe and
the central station to a compatible revision.
b. Downgrade the transmitter firmware to a revision
compatible with the mainframe and central station
2. If SpO
3. If changing the ECG PCB does not correct the
problem, replace the Main PCB.
TRANSMITTER MALF INOP
This INOP indicates that the self-test has discovered a
problem with the internal, non-RF circuitry of the
transmitter. The problem probably lies in the Digital ASIC
or the RAM section of the Main PCB. If this INOP
appears, replace the Main PCB.
ARRHY REQUIRED INOP
This message indicates that arrhythmia monitoring has
been turned off for an EASI ECG transmitter.
1. Turn arrhythmia monitoring ON at the Information
2. If arrhythmia monitoring is not desired, then a
standard ECG transmitter must be used.
Transmitter Non-RF Application
, cannot be monitored, replace the ECG