User 1: Appears when the monitor is operated by User 1.
User 2: Appears when the monitor is operated by User 2.
Weak Battery Symbol: Appears when batteries should be replaced.
Pulse Symbol: Shows the heart rate per minute.
Risk Category Index: See page 28 for more information.
Irregular Heartbeat Detector: See page 29 for more information.
HL858CB ( Bluetooth)
Data Transmitting Symbol: Appears when data is transmitting to your
LCD View Area: 74mm x 95 mm
• Rest Assure ON: The REST symbol will appear on monitor, then will countdown
for 5 minutes before starting single measurement.
• Triple Check mode ON: The REST symbol will flash among 3 measurements
Memory Average: Displays average of last 3 readings.
AM/PM Averaging: Indicates the reading being displayed is an average from
the last 3 morning or last 3 evening measurements.
Multi-Read Symbol: Appears when Triple Check Measurement System is
Average Symbol: Displayed when viewing a Triple Check average.
Measurement Indicator: Indicates which measurement is being taken, or
which measurement is being viewed from a Triple Check average reading.
and any of the following letters and numbers appear in the area
that systolic pressure should be displayed, an error has occurred with your
reading. See Troubleshooting section of this manual for more information.
Measurement Error: Wrap the cuff correctly and keep arm steady during
measurement. Take new measurement.
Air Circuit Abnormality: Check cuff connection. Take new measurement.
Pressure Exceeding 300 mmHg: Turn the unit off to clear, then
take new measurement.
Error Determining Measurement Data: Take new measurement.
Data Transmission Error: Monitor cannot connect to the mobile device to
transmit data. Make sure Bluetooth
System Error: Take new measurement.