Factory series electronic precision scales (213 pages)
Summary of Contents for Sartorius IB 16000 S
98648-000-60 Sartorius Industry IB 16000 S, IB 31000 P, IB 31. Electronic Precision Scales Installation and Operating Instructions...
Platform cover for scale base Tare control Level indicator (only for IB 16000S) CAL key Power receptacle ON/OFF key Leveling foot Weight display Menu access switch Manufacturer’s label (for scale operating program)
Contents Page About the Product (Warranty) Storage and Shipping Conditions Equipment Supplied Installations Instructions Ambient Conditions How to Mount the Display Unit Mounting the Display Unit on the Short Side of the Scale Base Mounting the Remote Display Unit Mounting the Display Unit on the YDH 01 B Support Arm (Option) Startup Connecting the scale to Line Power...
About the Product (Warranty) With this Sartorius Scale you have acquired a high-quality electronic Instrument that will ease your daily work load. Please read these installation and operating instructions carefully be- fore operating your new scale. Pursuant to the German Directive for the Implementation of Regulations for Prevention of Accidents "Elektrische Anlagen...
Do not miss out on the benefits of our full warranty. Please complete the warranty card, indicating the date of installation, and return the card to your Sartorius dealer. Unpack the scale. Remove the plastic covering and adhesive strips. Place the platform cover (1) on the scale base.
Equipment Supplied The equipment supplied includes the components shown on the left: Scale with platform and display unit AG adapter Dust cover 3 Allen wrenches Retainers for mounting the display 2 M 4 x 8 screws and washers (for mounting the "Remote Display Unit")
The components used in the scale are rated to at least Class KSF according to DIN 40040. Your Sartorius scale provides accurate readouts even when it is exposed to unfavorable ambient conditions. You can adapt the scale to your requirements simply by changing the code settings in the scale operating menu.
How to Mount the Display Unit The display unit can be mounted as follows: — base-mounted on the front of the scale (factory-mounted) — base-mounted on the side of the scale (see next page for direc- tions) — mounted as a remote display unit (see page 11) —...
Mounting the Display Unit on the Short Side of Scale Base Leave the platform cover on the scale base. To mount the display, turn the scale upside down so that it rests on the platform cover. Remove the screws from the base plate using the Allen wrench sup- plied (see arrows) and then remove the base plate.
Attach the display unit to the short side of the scale base by tightening the two screws. Insert the cable in the raceway as shown in the diagram on the left. Make sure that the rubber seal (see arrow) is not damaged. Reposition the base plate and fasten it in place by tightening the three M3x8 screws (see the large arrows) and the two M3x10 screws (see the small arrows).
Mounting the Remote Display Unit Leave the platform cover on the scale. To mount the display, turn the scale upside down so that it rests on the platform cover. Remove the screws from the base plate (see arrows using the Allen wrench supplied and then remove the base plate.
Attach the display unit to the mounting retainers supplied using the two M4x8 screws and washers (can also be mounted on a wall for use as a wall console). Thread the cable through one of the three openings in the scale hous- ing.
Mounting the Display Unit on the YDH 01IB Support Arm (Option) Leave the platform cover on the scale. To mount the display, turn the scale upside down so that it rests on the platform cover. Remove the screws from the base plate (see arrows using the Allen wrench supplied and then remove the base plate.
Remove the four screws from the display unit (see arrows) using the Allen wrench to untighten them. Unscrew the two screws from the cable clamp. Remove the clamp and then the display holder. Remove the cable from the raceway in the bottom of the scale.
Fasten the support arm to the rear panel of the scale base using the two M5x16 screws supplied and the star lock washers. Attach the mounting frame of the support arm to the bottom of the scale using the two M4x8 screws supplied. Refasten the rectangular cover plate to the support arm using the four screws.
Standard you use, please contact your Sarto- rius dealer. Important Note Only use original Sartorius AC adapters identified by the Sartorius la- bel. Use of AC adapters supplied by other manufacturers, even if these adapters have a registered approval rating from a national test- ing laboratory, requires the consent of a certified Sartorius technician.
Leveling the Scale using the Level Indicator (only for model IB 16000 S) At the point of use, level the scale using the leveling feet (4) so that the air bubble is centered within the circle of the level indicator (2).
Operation After you have plugged the scale into a wall outlet using the AC adapter, allow for at least 30 minutes' warmup. The weight display shows the following special Status messages for your Information: BUSY The scale processor is still busy processing a function and will not ac- cept any other commands to perform functions at this time.
Self-Test After the scale is turned on, an automatic self-test of the scale's elec- tronic circuitry is performed. This self-test ends with the readout 0.0 g/0.000 kg (if the factory code setting 5 1 1/ 5 1 2 is used). Weighing Now place your sample or object on the platform (1) to determine the weight.
Automatic Shutoff Make sure the "Automatic Shutoff" code is set to "ON" when using a non-rechargeable or rechargeable battery. This will increase the amount of time you can operate the scale before having to recharge the battery (see "Scale Operating Program"), lf you do not press a key or if a weight readout in the display does not change for at least 2 minutes, "STANDBY"...
Scale Operating Program The scale operating program lets you adapt your scale to various am- bient conditions and to different weighing requirements, and select weight units commonly used in your country. At the factory, we have set the codes for a Standard program which is protected by a locking function to prevent accidental changes.
After you have accessed the menu of the scale operating program, the display will show a continuous numerical sequence from 0 to 5 for the "page" or the first digit of the code, in addition to the Status code letter "L" or "C". When the first digit of the code you wish to check or change appears, press the tare control (6).
List of the Programmable menu Code Settings C 1 3 1 Page Line Word Menu of the Balance Operating Program (Active Parameters) Code Ambient Conditions Code Program Lock Very stable C 1 1 1 C 4 1 1 Stable C 1 1 2 C 4 1 2 Unstable C 1 1 3...
Tare before weighing before weighing The air bubble of the level Level scale indicator (2) is not within the circle(only for IB 16000 S) “POWER OFF” The operating voltage is too Connect the scale or the blinks in the display...
In this case, notify the Sartorius Service Center or International Tech- nical Support Group based in Goettingen, Germany. Only service technicians who are authorized by Sartorius and have access to the required manuals are allowed to perform maintenance and repair work on the equipment.
Accessories (Options) Data printer with YDP 02-0D date/time and statistics functions Print speed approx. lines/sec. Printer housing (W x D x H) in mm 150 x 138 x 43 in inches 5.5 x 5.4 x 1.7 Interface YDO 01 IB “Data Input”...
Specifications Model IB 16000 S IB 31000 P IB 31 Capacity 16,1 4/8/16/31 Readability 0,1/0,2/0,5/1 Maximum overload capacity Tare range (by subtraction) 16,1 Standard deviation ≤±0,05 ≤±0,1/0,1/0,25/1 ≤±0,5 Max. linearity deviation ≤±0,2 ≤±0,2/0,2/0,5/1 ≤±1 Stabilization time (typical) Adaption to ambient conditions...
All rights reserved. No part of this publication may be printed or translated in any form or by any means without the prior written permission of Sartorius AG. Sartorius AG reserves the right to make change to the technology, features, specification and design of the equipment without notice.