d. Attach loop to center of cable yoke (Point A). Route
cable around right-hand side of cable yoke (Point B), and pass
it in front of cable yoke to pulley C. Continue around pulley C
in a counterclockwise direction to sheave D.
counterclockwise turn around sheave D.
potentiometer (Option 003) follow this step. If not, go on to
step f. From sheave D start cable in center of sheave E. Make
two turns in a counterclockwise direction, passing cable
beneath itself after each turn.
through the slot on bottom edge of sheave to hole in the
sheave. Run cable through the hole, and under screw and
washer on top of sheave. Tighten screw. Pass cable through
slot on top edge of sheave. Continue around sheave two more
turns in a counterclockwise direction, passing each turn
beneath itself, and continue to pulley F.
f. Pull cable around pulley F in counterclockwise
direction, pass it in front and around left-hand edge of cable
yoke (Point G) and pass it to screw (Point H) on Y motor block.
g. Pass cable around screw (Point H) in clockwise
direction. Firmly pull on cable (to keep the cable taut), and
NOTE: There is a nut on the bottom of Nut H. Make sure
this is tight.
h. Check cable tension as described in paragraph 5-75.
FIGURE 5-11. RESTRINGING DIAGRAM
After second turn run cable
i. Clip off the excess wire.
5-45. ELECTRICAL MAINTENANCE
The Model 7035B requires minimum electrical
It is carefully aligned during manufacture.
However, if the recorder ever requires alignment, specific
adjustment procedures are detailed in this section. Section VII,
Troubleshooting, contains additional material.
5-48. PERFORMANCE TESTS
This instrument should meet the following Hewlett-
Packard performance standards to assure operation within
specifications. If this instrument fails to meet the following test
specifications, refer to adjustment procedures within this
section or Section VI, Troubleshooting.
5-51. TEST EQUIPMENT
The instruments and accessories required for
completing performance tests are listed in Table 5-1.