Disclaimer * Windows and MS-DOS are trademarks and/or registered trademarks of Microsoft Corporation in the United States and/or other countries. * VESA is a trademark of the Video Electronics Standard Association. * All other products and company names are trademarks of their respective owners.
CAN/CSA-C22.2 No. 60601-1:08 Windows is a registered trademark of Microsoft Corporation. NEC is a registered trademark of NEC Corporation. ErgoDesign is a registered trademark of NEC Display Solutions, Ltd. in Austria, Benelux, Denmark, France, Germany, Italy, Norway, Spain, Sweden, U.K.
Declaration Declaration of the Authorized Representative Declaration of the Authorized Representative Means of Conformity NEC Display Solutions Europe GmbH declares that the product listed is in conformity with the essential requirements and Device Classiﬁ cation: Class I, non-measuring function provisions of the Council Directive 93/42/EEC, including the Applicable Rules: Annex IX, Rules 1.4 (Section 1)
WARNING TO PREVENT FIRE OR SHOCK HAZARDS, DO NOT EXPOSE THIS UNIT TO RAIN OR MOISTURE. ALSO, DO NOT USE THIS UNIT’S POLARIZED PLUG WITH AN EXTENSION CORD RECEPTACLE OR OTHER OUTLETS UNLESS THE PRONGS CAN BE FULLY INSERTED. REFRAIN FROM OPENING THE CABINET AS THERE ARE HIGH VOLTAGE COMPONENTS INSIDE. REFER SERVICING TO QUALIFIED SERVICE PERSONNEL.
WARNING: To avoid risk of electric shock, this equipment must only be connected to a supply mains with protective earth. WARNING: No modiﬁ cation of this equipment is allowed. WARNING: For EMC details information, please contact NEC local authority. •...
NOTE: As with all personal display devices, NEC DISPLAY SOLUTIONS recommends using a moving screen saver at regular intervals whenever the screen is idle or turning off the monitor when not in use.
Contents Your new NEC monitor box* should contain the following: • MD211G5 monitor with tilt/swivel/pivot/height adjust stand • Power Cord • Video Signal Cable (DVI-D to DVI-D cable)* • DisplayPort cable • USB Cable • Quick Reference Guide Power Cord*...
Quick Start To connect the LCD monitor to your system, follow these instructions: NOTE: Make sure to read “Recommended Use” before installation. In order to display the maximum resolution, a display controller that can output a resolution of 2048 x 2560 is needed. 1.
5. To keep the cables neatly organized, place them into the cable management system that is built into the stand. Place the cables in the hooks ﬁ rmly and evenly (Figure D). Figure D 6. Slide down the cable cover (Figure E). 7.
Controls 1 Power Switch To turn the monitor on and off. 2 LED Indicates that the power is on. Turns off automatically, when the luminance is stabilized. 3 Front Sensor For luminance stabilization and calibration. 4 Ambient Light Sensor Measures ambient light during calibration process. 5 + (plus) Button To open the OSD (On Screen Display) click –...
Speciﬁ cations Monitor Speciﬁ cations MD211G5 Notes LCD panel color Clear Base 21.3-inch (54 cm), monochrome, TFT active matrix, anti-glare Pixel pitch (H x V) 0.165 x 0.165 mm Display area (H x V) 337.9 x 422.4 mm Pixel (H x V) 2048 x 2560 Grayscale 256 steps of gray out of a palette of 12277...
Additional Installation Information Advanced OSD Several control functions are available by entering the advanced OSD. These control functions are additional OSD menu items, which are always accessible after opening the standard OSD. Control Function Description EDID Change the picture format to be recognized by the operating system •...
3. After selecting your preferred EDID mode, hold down the (+) plus button and the (-) minus button simultaneously for over 2 seconds again. Then displayed (Information) OSD and Selected (Landscape) EDID data is displayed as (FIXED). Then conﬁ rm the EDID line (Portrait/Landscape) on the “INFORMATION” OSD. 4.
How to Change Conﬁ guration (Preset 1, 2, 3) The NEC model MD211G5 is shipped with 3 factory preset conﬁ gurations, as listed below. Luminance (cd/m Conﬁ guration Gamma Remark Maximum Minimum Ambient DICOM [Factory Default] DICOM DICOM The conﬁ guration can be changed via an OSD menu. The conﬁ guration 1 parameters may be modiﬁ ed anytime by a new display calibration.
Performing the DICOM conformance test This function is available only when the display has been calibrated to DICOM GSDF. Performing the DICOM conformance test ensures DICOM conformance of the display by measuring and checking the display properties. The DICOM conformance test is recommended to be performed after 60 minutes of continuous lighting time of the backlight of the LCD panel.
4. Showing the test results When the measurement is completed, the test results are shown. Test results Test results Passed: Max. deviation ≤ 10% Failed: Max. deviation > 10% Canceled: Test has been canceled. Passed(!), Failed(!) “(!)” following Passed and Failed indicates that the recommended condition has not been met. (The test was performed in less than 60 minutes of continuous lighting time of the backlight.) If the test is continuously performed in less than 30 seconds, “Unstable”...
Attaching a wall mount or multi-display mounting arm The tilt and swivel stand, as shipped with the display, is interchangeable with commercially available wall mounts or multi-display mounting arms that: • Come with a mount plate with 4 slotted screw holes at 100 mm x 100 mm and •...
Dedicated recycling sites ensure that environmentally harmful components are properly handled and securely disposed. To ensure the best recycling of our products, NEC DISPLAY SOLUTIONS offers a variety of recycling procedures and gives advice on how to handle the product in an environmentally sensitive way, once it has reached the end of its life.
NEC Display Solutions, Ltd. NEC Display Solutions of America NEC Display Solutions Europe GmbH 4-28, Mita 1-chome, Minato-ku, 500 Park Blvd. Suite 1100 Itasca, Landshuter Allee 12-14 Tokyo, Japan Illinois 60143 D-80637 Muenchen Germany Phone: +1-630.467.3000 Phone: +49(0)89/99699-0 Fax: +1-630.467.3010 Fax: +49(0)89/99699-500 Vol.