Medical Imaging The NEC MD21GS-2MP is designed for 2-megapixel (1200 x 1600) grayscale medical imaging. This display comes fully tuned with a setting for gamma correction that complies with the DICOM Part 14 Standard. Canadian Department of Communications Compliance Statement DOC: This Class B digital apparatus meets all requirements of the Canadian Interference-Causing Equipment Regulations.
Contents Your new NEC Display display box* should contain the following: • MD21GS-2MP Display with tilt/swivel/pivot/height adjustable stand • Power Cords (N.A. Hospital Grade, European Continental and Chinese) • Quick Reference Guide • Installation and Maintenance Guide • Video Signal Cable (DVI-D to DVI-D cable) •...
Quick Start Display Card* Installation 1. Following your PC manufacturer’s guidelines, open your computer to access the PCI, AGP or PCI E slot. 2. Place the display card into an available PCI, AGP or PCI E slot and secure all screws. 3.
Quick Start –continued 4. For digital input: Connect the DVI-D signal cable to the connector on the back of the display (Figure C.1). For analog input: Connect the 15-pin mini D-SUB signal cable (not included) to the connector on the back of the display (Figure C.2).
User Controls The On-Screen Manager (OSM) control buttons are located on the back of the display. To access the OSM menu, press any of the user control buttons ( < , >, –, +). 1. LED Indicates power on and signal condition. The LED can be turned on, off, or dimmed. Please refer to the Advanced OSM section/LED Brightness for further information.
To alleviate image persistence, turn off the monitor for as long as the previous image was displayed. For example, if an image was on the monitor for one hour and a residual image remains, the monitor should be turned off for one hour to erase the image.
(LCD); 0.270mm dot pitch; 400 cd/m calibrated white luminance; 700:1 contrast ratio, typical. VideoDot Clock 162MHz Max When used with an NEC recommended 10 bit display card When used with an NEC recommended 8 bit display card Automatically...
References North America and Canada NEC-Mitsubishi Monitor Customer Service & Support Customer Service and Technical Support: Parts and Accessories/Macintosh Cable Adapter: Warranty Information: Online Technical Support Sales and Product Information Sales Information Line: Canadian Customers: Government Sales: Government Sales email:...
FCC Rules. OSM and FullScan are registered trademarks of NEC-Mitsubishi Electronics Display of America in U.S. X-Light is a trademark of NEC- Mitsubishi Electronics Display of America in U.S. ErgoDesign is a registered trademark of NEC-Mitsubishi Electric Visual Systems Corporation in Austria, Netherlands, Denmark, France, Germany, Italy, Republic of Indonesia, Republic of Korea, Kingdom of Thailand, Malaysia, Norway, Spain, Taiwan, U.K., U.S., Canada.