Designed for in vitro diagnostic use only, testing requires just a small amount of fresh capillary whole blood from either the fingertip, palm or forearm. If you have any questions regarding the GE100 Blood Glucose Monitoring System or testing procedures, contact our Customer Support staff or your healthcare professional.
Blood Glucose Monitoring System has not been used for arterial blood testing yet, so it is not suitable to be used for arterial blood testing. CAUTION If the GE100 Blood Glucose Meter and Test Strips are exposed to a substantial change in temperature, - The GE100 Blood Glucose Meter will be set to mg/dL by default when sold in the United States.
Table of Contents Table of Contents About The GE100 System Quality Control GE100 Blood Glucose Monitoring System...........................8 Quality Control Test.....................................34 GE100 Blood Glucose Meter..............................10 Performing a Quality Control Test..............................36 GE100 Blood Glucose Test Strip............................14 Test Memory Before Testing Recalling Test Results and Average..............................38 Meter Activation and Battery change............................16...
Blood Glucose Monitoring System GE100 Blood Glucose Monitoring System 3 ＊ Your GE100 Blood Glucose Monitoring System consists of several items. Please identify each item, learn GE100 its name and how it is used. 4 ＊ Below are the items included in your GE100 Blood Glucose Monitoring System: 1.
Battery Cover Display Window Product name Meter Battery Meter Serial Number Main Button USB Port (For data downloading on authorized GE software only) NOTE: Do not perform a blood glucose test when USB cable is connected to the Blood Glucose Meter.
The GE100 Blood Glucose Meter The GE100 Blood Glucose Meter Indicates a test result stored in memory Indicates when to apply the blood sample Indicates a test result not included into the Average Test strip function of this meter Occurs when strip is not compatible with your meter Indicates the average result or if a strip is inserted incorrectly.
Electrochemical sensor Do not use expired GE100 Blood Glucose Test Strips . - When you open a new vial of GE100 Test Strips, record the date on the vial. - Discard the vial of test strips after 3 months from opening.
2. With Battery Inserted Press the main button first to turn on the GE100 Blood Glucose Meter. Hold down the main button for 7 Seconds. During this time the screen will go blank and then after hearing a beep it will turn on into Setting Mode.
Setting Up Your Meter- Setting the Time, Volume and Date Setting Up Your Meter- Setting the Time, Volume and Date 1. Year setting 4. Time format 12/24H selection With the time format blinking, press the Up or Down With the year format blinking, press the Up or Down button until you see the button to switch between 12H and 24H.
2. Manual Power off If you want to turn off the GE100 Blood Glucose Meter, press and hold the Main button for 3 seconds. 3. Auto Power off GE100 Blood Glucose Meter will power off automatically after 2 minutes if no buttons are pressed or strip is inserted.
GE100 3. Take the GE100 Blood Glucose Test Strip out of the test strip port. Please follow 3. Insert the GE100 Blood Glucose Test Strip into test strip port until it clicks and your local regulations and discard used strips properly.
Bend the cap down. When a gap appears between the cap and base, GE100 pull them off in opposite directions. - GE100 Blood Glucose Meter - GE100 Blood Glucose Test Strips (Please check expiration date of test strips. Do not used expired test strips) 2. Pull off the depth adjustable cap.
(Refer to 10. Insert the strip into the Test Strip Port of the GE100 Blood page 5 for Sample Size Chart.) Discard the test strip and Glucose Meter with the indication symbol facing up. The repeat with a new GE100 Blood Glucose Test Strip.
- To do the alternative site testing, install the clear cap for your GE100 Lancing Device. (For detailed information check the instruction manual for the GE Lancing 16. Hold the release button in one hand (see picture 16, step 1) Device.)
GE100 Blood Glucose Test Strips. maintained and monitored by a healthcare provider. - Use each GE100 Blood Glucose Test Strip immediately after taking it from the vial. - Do not reuse GE100 Blood Glucose Test Strips. - Do not perform testing immediately after moving from one ambient temperature to another (e.g.
GE100 Blood Glucose Test Strip. You can also run a Quality Control Test with the GE100 Control repeat your test again with a new strip. Solutions to check your GE100 Blood Glucose Meter and Test Strip ( Refer to page 36 ). If the test result still If you still get "...
Control Solution Range: - Your GE100 Blood Glucose Test Strips have expired. - You left the cap of the GE100 Blood Glucose Test Strip vial or the GE100 Control Solution off for a Example of Control Solution Range Test Strips Control Solution Range mg/dL period of time.
" and be accompanied by beeps (if volume is turned on). - Don't drip the GE100 Control Solution into the sample port of the strip directly. The reagent on 7. Gently touch the sample entry of the strip with the GE100 Control the strip might leak into the bottle of GE100 Control Solution and may cause Solution from the top of the cap.
Recalling Test Results and Average Recalling Test Results and Average The GE100 Blood Glucose Meter is able to automatically store a maximum of 500 test results with time and date. If your meter has stored 500 results, the newest test result will replace the oldest one. To recall your test memory, start the meter without a test strip inserted.
Up or Down button for 2 seconds. (Down button is for searching the latest test value to the oldest one; Keep your GE100 Blood Glucose Meter and Test Strip free of dust, water or any other liquid. Store the Up button scrolls from the oldest to the latest one).
Quality Control Test (see page 36-Performing the battery immediately. a Quality Control Test). Er3 - Signal transmission is disrupted, repeat the test. If the error messages continue after attempting to remedy, please contact GE Diabetes Customer Support Center by calling: 1-866-613-7085.
Error Messages and Troubleshooting Sampling Error Temperature Error Do not apply the blood drop to the sample entry of the strip before the GE100 Blood In order to get an accurate test result, perform testing between 50~104°F 10~40°C Glucose Meter displays "...
°C to 30 °C ) and relative humidity below 90%. Do not freeze. - GE100 Blood Glucose Test Strips are designed for using with capillary whole blood samples. Do not use Measurement Calibration Plasma serum or plasma samples. - Inaccurate test results may be obtained at altitudes greater than 10,000 feet (30048 meters) above sea...
1 CR2032 battery Power Supply This warranty does not apply to the performance of a GE100 Blood Glucose Meter that has been altered, misused, tampered with or abused in any way. Meter Battery Life About 1000 tests This warranty applies only to the original purchaser of the GE100 Blood Glucose Meter.
Customer Service Expected glucose values without diabetes Please review all the instructions to make sure you are performing the steps correctly. If you have any Expected glucose values without diabetes questions or concerns, please contact us toll free at 1-866-613-7085 (Monday through Friday 8:00 AM Range (mg/dL) to 5:00 PM PST) or visit our web site: www.gediabetes.com.
Essential Components List Log Book GE100 Blood Glucose Meter, Test Strip and Control Solution Manufacturer: Bionime Corporation Name: Product complies with In Vitro Diagnostic Medical Device Directive 98/79/EC ( CE0197 ) Address: GE100 Lancing Device Manufacturer: Bionime Corporation Home Phone:...
2. Guarantees the performance of the GE100 Blood Glucose Meter only if used as directed and provided that the failure to find me in a coma or stupor, perform has not been caused in whole or in part by the use of test strips that are not GE100 Test Strips. Use only GE100 Blood • Doctor/Hospital: p l e a s e c o n t a c t n e a r e s t Glucose Test Strips in your GE100 Blood Glucose Meter.