Distribution and Customer Service in Canada by General Electric Company and are used Auto Control Medical The GE200 Blood Glucose Monitoring System is designed for in vitro diagnostic use only (for single patient use and under license by Bionime Corporation, NO. www.autocontrol.com self-testing only).
The GE200 Blood Glucose Monitoring System is intended to be used for the quantitative measurement of glucose in - The GE200 Blood Glucose Meter can only be used with GE200 Blood Glucose Test Strips. Other brands’ test strips fresh capillary whole blood samples drawn from the fingertips, forearm or palm. The GE200 Blood Glucose should not be used under any circumstances.
- The minimum blood sample size to test using the GE200 Blood Glucose Monitoring System is 0.75µL: ( - The GE200 Blood Glucose Meter and Lancing Device are for single patient use. Do not share them with anyone including other family members.
Turning On/Off the Meter..........................................22 Cleaning and Disinfecting Procedures.....................................47 Turning On/Off the Backlight........................................23 Error Messages and Troubleshooting.......................................51 Handling the GE200 Blood Glucose Test Strip..................................24 Limitations................................................55 Specifications...............................................56 Warranty................................................58 Testing Procedure Customer Service.................................
The GE200 Blood Glucose Monitoring System 3 ＊ Contents Your GE200 Blood Glucose Monitoring System consists of several items. Please identify each item, learn its name and how it is used. 4 ＊ Below are the items included in your GE200 Blood Glucose Monitoring System: 1.
The GE200 Blood Glucose Meter The GE200 Blood Glucose Meter Test Strip Port Product name Meter Serial Number Display Window Meter Battery Main Button ( Battery Cover Backlight Battery Left Button ( Right Button (...
The GE200 Blood Glucose Meter The GE200 Blood Glucose Meter Indicates a test result stored in memory Indicates when to apply the blood sample Indicates a test result not included into the average Test strip results of the meter Blinks when strip is not compatible with your meter or if a Indicates the average of the test results strip is inserted incorrectly.
- Do not reuse GE200 Blood Glucose Test Strips. - Do not use expired GE200 Blood Glucose Test Strips. - When you open a new vial of GE200 Blood Glucose Test Strips, record the date on the vial. Discard the vial Noble Metal Electrodes Hold here to of test strips after 4 months from opening.
GE200 Blood Glucose Meter comes with two CR2032, 3volt, batteries installed. Two new batteries will provide 4. The GE200 Blood Glucose Meter will enter Self-Testing Mode automatically when replacing the battery (all symbols power to perform approximately 1000 tests under normal use. Press the main button or insert a strip to activate your will appear on the screen).
With the month blinking, press the Left or Right button until you see the current month. Press Press the main button to turn on the GE200 Blood Glucose Meter. Hold down the main button for 7 seconds. During the Main button to confirm. Once the month is confirmed, the Day Setting will appear.
With the minutes blinking, press the Left or Right button until you see If you do no change any settings during Meter Set-up for over 2 minutes, the GE200 Blood Glucose Meter the current minute. Press the Main button to confirm. Once the will leave setting mode and power off automatically.
The GE200 Blood Glucose Meter backlight will turn off automatically after 10 seconds if no buttons are pressed. The GE200 Blood Glucose Meter will power off automatically after 2 minutes if no buttons are pressed or no strip is inserted.
2. Put your forefinger on the side of the strip as shown. 3. Insert the GE200 Blood Glucose Test Strip into test strip port until it clicks and firmly stops. 3. Take the GE200 Blood Glucose Test Strip out of the test strip port.
Bend the cap down. When a gap appears between the cap and base, pull the cap and base in opposite directions. - GE200 Blood Glucose Meter - GE200 Blood Glucose Test Strips 2. Pull off the depth adjustable cap. (Please check the expiration date on the test strip vial. Do not use expired test strips) 3.
14. Touch and hold the blood drop to the edge of sample port until the view window 10. Insert the strip into the test strip port of the GE200 Blood Glucose Meter with the view window facing up. is filled with blood. If the view window is not completely filled with blood the test 11.
17, step 2) to safely eject the used disposable - The GE200 Blood Glucose Monitoring System and GE Lancing Device are intended for single lancet. patient use only. Do not share these devices with anyone, including other family members.
Palm CAUTION - To perform alternative site testing, install the clear cap for your GE Lancing Device. - Test results may vary if blood samples are taken from different sites or under certain conditions where ( For detailed information, refer to the instruction manual for the GE Lancing Device. ) glucose levels are changing rapidly such as: following a drink, a meal, an insulin dose or exercise.
GE200 Test Strip (Refer to “Performing a Quality Control Test” on page 40) . CAUTION - Check the expiration date printed on the strip vial every time you use a test strip. Do not use expired GE200 If the test result still remains unusually high or low, contact your healthcare professional immediately.
If the test result is below 1.1 mmol/L, “ ” will appear on the screen. Please repeat your test Solution Mode. If the test result is within the Control Solution Range printed on the strip vial label, the GE200 Blood with by a new test strip.
- To ensure that your GE200 Blood Glucose Meter and GE200 Blood Glucose Test Strip are working properly. - Each time you open a new bottle of control solution, write the expiration date on the label. The GE200 - To confirm that you are following the correct testing procedures.
Your control solution test results should fall within the control solution range. If the results are within the range, the The GE200 Blood Glucose Meter is able to automatically store a maximum of 1000 test results with time and date. If GE200 Blood Glucose Monitoring System is working correctly.
Recalling Average Test Results The GE200 Blood Glucose meter provides several average test results. View the 1-day, 7-days, 14-days, 30-days, 60- To recover excluded test results and include them in the average calculation, do not leave the current mode. Press the Right button until the symbol "...
If your test strip port is stained with blood, control solution or any liquid, please use a dry tissue or alcohol swab to clean it immediately. Do not use anything wet to clean it. Perform a quality control test to ensure the GE200 Blood CaviWipes Disinfecting Towelettes has been tested and may be used to clean and disinfect the meter and lancing Glucose Meter is working properly.
- Your GE Lancing Device has been tested to ensure that there is no change in the performance of the NOTE - Your GE200 Blood Glucose Meter has been tested to ensure that there is no change in the performance or device after550 cleaning cycles and 550 disinfecting cycles. The testing simulates 2 cleaning and external materials of the device after 550 cleaning cycles and 550 disinfecting cycles.
- You are unable to turn on/off your meter, operate the right / left button, the lancing device 3. If the GE200 Blood Glucose Meter and Test Strips are exposed to a substantial change in release button or depth adjustable cap.
If the GE200 Blood Glucose Meter will not turn on, please follow the steps below: you may have used the wrong test strips. Please check the test strip vial to ensure you are using the GE200 Test strips. 1. Open the battery cover and remove the batteries.
Error Messages and Troubleshooting Limitations - GE200 Blood Glucose Monitoring System is not intended for serum or plasma testing. CAUTION - Inaccurate test results may be obtained at altitudes greater than 10,000 feet (3,048 meters) above sea level. A blood sample should ONLY be applied to the test strip after the test strip has been inserted correctly and the Meter's screen is showing the image of the test strip and a blood drop flashing symbol.
Specification Specification Measurement Technology Dehydrogenase Electrochemical Sensor Hematocrit 10-70% Sample Capillary whole blood Power Supply Two CR2032 Batteries Minimum Sample Volume 0.75 microliters Battery Life Approximately 1000 standard tests Meter Dimension Measuring Range 1.1 - 33.3 mmol/L 96 mm x 46 mm x 17.5 mm Meter Weight Test Time 5 seconds...
Auto-Control Medical Distribution and Customer Service Center at 1-800 964 8434 . We This warranty does not apply to the performance of a GE200 Blood Glucose Meter that has been altered, misused, will make every effort to assist you.
Expected Glucose Values Without Diabetes Component Manufacturer Information Expected glucose values without diabetes ⁽¹⁾ Blood Glucose Meter, Test Strip, Control Solution Manufacturer: Bionime Corporation Range (mmol/L) Status NO. 100, Sec. 2, Daqing Street, South District, Taichung City, Taiwan 40242 Product complies with In Vitro Diagnostic Medical Device Directive 98/79/EC. ( CE0197 ) <5.6 Fasting EC-Rep: Bionime GmbH, Klaus Ellensohn, Tschuetschgasse 8, 6833 Klaus / Vlbg., Austria...
Or call: perform has not been caused in whole or in part by the use of test strips that are not GE200 Blood Glucose Test Strips manufactured by Bionime. Use only GE200 Blood Glucose Test Strips in your GE200 Blood Glucose Meter.