USER GUIDE ReceiveR-in-eaR HeaRing instRument...
Intended use Hearing instrument type designations for models included in this user guide are: SY312, FCC ID: X26SY312, IC: 6941C-SY312; SY312e, FCC ID: X26SY312e, IC: 6941C-SY312e; and MRIE, FCC ID: X26MRIE, IC: Generic air-conduction hearing instruments are wearable sound-amplifying devices intended to compen- 6941C-MRIE.
Practice and a positive attitude are important in learning to use hearing instruments. Your ReSound instruments have been adjusted according to your individual hearing loss and needs. Some people adjust quickly to wearing hearing instruments in Left serial number: their ears and hearing new sounds;...
312 battery are available in the The identification number for the MRIE, SY312 and Flight mode ....... . 17 Prescription use of this TSG hearing instrument .
1. Receiver tube (Indentification number for the 62 models 61 models 2. Receiver Open Dome SY312e and SY312 devices) (type MRIE) (type SY312e 9, 10 3. Receiver Tulip Dome 60 models and SY312) 4. Receiver Power Dome 5. RIE mould 6.
After this, your hearing instrument and ReSound Unite Remote Control will continue to work as usual, hearing instrument to allow activation of the battery. but you will not be able to use your ReSound Unite TV Streamer, Phone Clip and Mini Microphone. At some 3. Insert the new battery with the positive side in the point the battery level will not support the remote control either and you will once again hear the descending correct position.
Sports lock Insertion (domes) 1. Hold the receiver tube where it bends, and The sports lock will be applied or adjusted by your hearing care professional. gently place the dome into the ear canal. Push Inserting/Removing hearing instruments the dome far enough into the ear canal so that the receiver tube lies flush with the side of the Insertion (custom RIE moulds) head.
Program and Multi-Function buttons When using the program or multi-function buttons to switch programs, each press will move the instrument to the next program. For example, if it was in program 1 it will switch to program 2, if it was in program 2 it If you have a hearing aid with a program button or multi-function button, this will allow you to use up to four will switch to program 3 etc.
If necessary, your hearing care practitioner can change these settings and fill in the following table to indi- Flight mode* (Wireless models only) cate the new settings: When boarding a flight or entering an area where RF transmitters are prohibited, wireless functionality must be deactivated, as it is not allowed to radiate radio signals during flights or in otherwise restricted areas.
Telephone use Note: It is important to wait an additional 15 seconds after wireless function resumes before open- If your hearing instruments are fit with a receiver open dome or receiver tulip dome, you can probably ing and closing the battery compartment again for any reason. If the battery compartment is opened and use the telephone as you normally would by holding it up to your ear canal opening.
Listen to radio or TV 1. Clean the telephone receiver thoroughly. 2. Hold the telephone vertically, in a position similar to when making a When listening to the TV or the radio, start out by listening to news commentators since they usually speak telephone call.
The sound source is connected to the hearing instruments by a cable or a wireless FM system to the audio receiver. boot. This accessory connects to the bottom of the hearing instruments, and once properly clicked into 4. Only use magnets supplied by ReSound. place, the hearing instruments switch to DAI automatically. Telecoil (optional)
3. Avoid rough handling of hearing instruments or dropping them on hard surfaces or floors. 2. Once in place, move the boot in the direction of the 4. Do not leave hearing instruments in or near direct heat or sunlight, such as in a hot, parked car, as battery compartment.
The receiver tube Changing wax guard for receiver tube The receiver tube contains the wiring to the receiver which delivers the sound to the ear canal. It is important that the receiver tube and the receiver dome/RIE mold fits correctly in your ear. If the receiver tube or the receiver dome/RIE mould irritates your ear in any way and prevents you from wearing your hear ing instru- ment, please contact your hearing care pro fessional.
Intended use of smart phone apps: GN ReSound smart phone apps are intended to be used with GN ReSound wireless hearing aids. GN Re- Sound smart phone apps send and receive signals from the GN ReSound wireless hearing aids via smart phones for which the apps have been developed.
* ReSound wireless devices operate in the frequency range of 2.4 GHz - 2.48 GHz. General warnings * ReSound wireless devices include a RF transmitter that operates in the range of 2.4 GHz - 2.48 GHz. * For use of wireless functionality only use ReSound Unite accessories. For further guidance regarding e.g.
Your doctor, audiologist or hearing healthcare Your ReSound hearing instruments may also include the Tinnitus Sound Generator function, a tool for professional will review with you the need for having such a control.
The scientific concepts that form the basis for the device Prescription use of this TSG hearing instrument The TSG Module provides sound enrichment with the aim of surrounding the tinnitus sound with a neutral The TSG module should be used as prescribed by your doctor, audiologist or hearing healthcare profes- sound which is easily ignored.
a tool to generate sounds to be used with appropriate counselling and/or in a tinnitus management program Battery warning information to relieve patients suffering from tinnitus. Batteries, although very small, contain dangerous substances, and should be disposed of carefully. This is for the safety of you and the environment.
Following the medical evaluation, the physician will give you a written statement that states that your hear- Warning to hearing aid dispensers (US Only) ing loss has been medically evaluated and that you may be considered a candidate for a hearing aid. The A hearing aid dispenser should advise a prospective hearing aid user to consult promptly with a licensed physician will refer you to an audiologist or a hearing aid dispenser, as appropriate, for a hearing aid evalu- physician (preferably an ear specialist) before dispensing a hearing aid if the hearing aid dispenser de-...
Troubleshooting Guide Symptom cauSe poSSIBLe RemeDy No sound Not turned on Turn on by closing the battery door Dead battery Replace battery Battery door will not close Insert battery properly Blocked RIE mould or tube Clean RIE mould or tube Blocked wax filter Replace wax filter or consult your hearing care professional Not loud enough...
Troubleshooting Guide Symptom cauSe poSSIBLe RemeDy Excessive whistling / Incorrect RIE mould placement Re-insert RIE mould carefully feedback Incorrect dome placement Re-insert dome Excessive ear wax Consult your hearing care professional Feedback control may need adjustment Consult your hearing care professional RIE mould tubing worn or damaged Consult your hearing care professional Hearing instrument settings not optimal...
Technical data Micro and Mini RIE—HP receiver Micro and Mini RIE—NP receiver MAXIMUM OUTPUT (OSPL 90) Models: AL960-DR, AL760-DR, AL560-DR, AL961-DRW, AL761- MAXIMUM OUTPUT (OSPL 90) Models: AL960-DR, AL760-DR, AL560-DR, AL961-DRW, DRW, AL561-DRW, ALT960-DR, ALT961-DRW 2cc Coupler 2cc Coupler 2cc Coupler AL761-DRW, AL561-DRW, ALT960-DR, ALT961-DRW Open Closed...
RIE—NP receiver RIE—HP receiver MAXIMUM OUTPUT (OSPL 90) MAXIMUM OUTPUT (OSPL 90) 2cc Coupler 2cc Coupler 2cc Coupler Models: AL962-DVIRW, AL762-DVIRW, AL562-DVIRW, Models: AL962-DVIRW, AL762-DVIRW, AL562-DVIRW, ALT962-DVIRW ALT962-DVIRW Open Closed Open Closed Reference test gain (60 dB SPL input) Reference test gain (60 dB SPL input) Full-on gain (50 dB SPL Input) Full-on gain (50 dB SPL Input) Maximum output (90 dB SPL input)
Micro and Mini RIE—S receiver Micro and Mini RIE—NP receiver MAXIMUM OUTPUT (OSPL 90) MAXIMUM OUTPUT (OSPL 90) Models: VO960-DR, VO760-DR, VO560-DR, 2cc Coupler Models: VO960-DR, VO760-DR, VO560-DR, 2cc Coupler VO961-DRW, VO761-DRW, VO561-DRW VO961-DRW, VO761-DRW, VO561-DRW VOT960-DR, VOT760-DR, VOT961-DRW, VOT761-DRW VOT960-DR, VOT760-DR, VOT961-DRW, VOT761-DRW Open Closed...
Micro and Mini RIE—HP receiver RIE—S receiver MAXIMUM OUTPUT (OSPL 90) MAXIMUM OUTPUT (OSPL 90) Models: VO960-DR, VO760-DR, VO560-DR, 2cc Coupler Models: VO962-DRW, VO762-DRW, VO562-DRW 2cc Coupler VO961-DRW, VO761-DRW, VO561-DRW VOT962-DRW, VOT762-DRW VOT960-DR, VOT760-DR, VOT961-DRW, VOT761-DRW Open Closed Reference test gain (60 dB SPL input) Open Closed Full-on gain (50 dB SPL Input)
RIE—NP receiver RIE—HP receiver MAXIMUM OUTPUT (OSPL 90) MAXIMUM OUTPUT (OSPL 90) Models: VO962-DRW, VO762-DRW, VO562-DRW 2cc Coupler Models: VO962-DRW, VO762-DRW, VO562-DRW 2cc Coupler VOT962-DRW, VOT762-DRW VOT962-DRW, VOT762-DRW Open Closed Open Closed Reference test gain (60 dB SPL input) Reference test gain (60 dB SPL input) Full-on gain (50 dB SPL Input) Full-on gain (50 dB SPL Input) Maximum output (90 dB SPL input)
Warranty and repairs Be aware of information marked with the warning symbol ReSound provides a warranty on hearing instruments in the event of defects in workmanship or materials, WARNING points out a situation that could lead to serious injuries, as described in applicable warranty documentation. In its service policy, ReSound pledges to secure func- CAUTION indicates a situation that could lead to minor and moderate injuries”...
Fax: (free) 0800 007 695 Fax: +44 1869 343 466 www.gnresound.com.au www.gnresound.co.nz CVR no. 55082715 www.gnresound.co.uk Any issues relating to the EU Medical Device Directive 93/42/EEC, or Council Directive 1999/5/EC on Radio Equipment and Tele communications terminal equipment should be directed to ReSound A/S...