- Measure-during-inflation technology - Supports 2 users Intended use The Philips upper arm blood pressure monitor is a digital monitor intended for use in measuring blood pressure and heart rate in adult patient population with arm circumference ranging from 22cm to 42cm (about 9-17 inches).
–20ºC device can be safely exposed: -20°C to 60°C. Symbol for the 2 year Philips guarantee. The Green Dot ('Der Grüne Punkt' in German) is the license symbol of a European network of industry-funded systems for recycling the packaging materials of consumer goods.
- Do not open, disassemble or repair the device yourself. Modification of the device voids your warranty. - Report to Philips Consumer Care if any unexpected operation or events occur. - Dispose of accessories, detachable parts, and the ME equipment according to the local guidelines.
Display Sy Description Explanantion m Systolic Maximum blood pressure, blood see also section systolic and pressure diastolic pressure. Diastolic Minimum blood pressure, see blood also section systolic and pressure diastolic pressure. Heart rate Number of heartbeats per minute (pulse is typically equivalent to heart rate).
Google Play. Use the search term 'Philips HealthSuite health app'. The app is available for iOS 8.0+ and Android 4.4+. Note: You can only use the Philips HealthSuite health app to communicate with the device. It is not possible to use third party applications.
Transmit and store personal health data in the app Note: Your personal measurement data is only stored and displayed in the Philips HealthSuite health app. Activate the Philips HealthSuite health app and Bluetooth on your mobile device directly after a measurement.
Note: We strongly advise you to take your product to an official collection point or a Philips service centre to have a professional remove the battery. Caution: Observe basic safety precautions when you follow the procedure described below.
The display A system Retake the shows EExx error measurement. If occurred. the problem persists, contact the Philips Consumer Care Center in your country. Data Bluetooth is Turn on Bluetooth transmission off. on your mobile or pairing device.
The device is approved according to EMC safety standard EN 60601-1-2. It is designed to be used in typical domestic environments. Hereby, Philips Consumer Lifestyle BV declares that the radio equipment type Bluetooth LE is in compliance with Directive 2014/53/EU.
Table 1 Guidance and manufacturer's declaration – electromagnetic emissions - for all ME equipment and ME systems Guidance and manufacturer’s declaration – electromagnetic emissions The device is intended for use in the electromagnetic environment specified below. The customer or the user of the device should assure that it is used in such an environment.
the equation applicable to the frequency of the transmitter. Recommended separation distance: d = 1.2 ÖP d = 1.2 ÖP 80 MHz to 800MHz d = 2.3 ÖP 800 MHz to 2.5 GHz where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer and d is the recommended separation distance in metres (m).