DL8760 EN-UK Specifications are subject to change without notice. © 2016 Koninklijke Philips N.V. All rights reserved Philips Consumer Lifestyle BV Tussendiepen 4, 9206AD Drachten, Netherlands Fax +31 (0)512594316 4222.100.4492.3 (6/2016) >75% recycled paper >75% papier recyclé 2 - 3 cm...
- Measure-during-inflation technology - Supports 2 users Intended use The Philips upper arm blood pressure monitor is a digital monitor intended for use in measuring blood pressure and heart rate in adult patient population with arm circumference ranging from 22cm to 42cm (about 9-17 inches).
–20ºC device can be safely exposed: -20°C to 60°C. Symbol for the 2 year Philips guarantee. The Green Dot ('Der Grüne Punkt' in German) is the license symbol of a European network of industry-funded systems for recycling the packaging materials of consumer goods.
- Do not open, disassemble or repair the device yourself. Modification of the device voids your warranty. - Report to Philips Consumer Care if any unexpected operation or events occur. - Dispose of accessories, detachable parts, and the ME equipment according to the local guidelines.
Sy Description Explanantion m Systolic Maximum blood pressure, blood see also section systolic and pressure diastolic pressure. Diastolic Minimum blood pressure, see blood also section systolic and pressure diastolic pressure. Heart rate Number of heartbeats per minute (pulse is typically equivalent to heart rate).
Bluetooth Smart function. Download the Philips HealthSuite health app from the App store or Google Play. Use the search term 'Philips HealthSuite health app'. The app is available for iOS 8.0+ and Android 4.4+.
Transmit and store personal health data in the app Note: Your personal measurement data is only stored and displayed in the Philips HealthSuite health app. Activate the Philips HealthSuite health app and Bluetooth on your mobile device directly after a measurement.
Note: We strongly advise you to take your product to an official collection point or a Philips service centre to have a professional remove the battery. Caution: Observe basic safety precautions when you follow the procedure described below.
The display A system Retake the shows EExx error measurement. If occurred. the problem persists, contact the Philips Consumer Care Center in your country. Data Bluetooth is Turn on Bluetooth transmission off. on your mobile or pairing device.
Problem Possible Solution cause The blood Place your mobile pressure device closer to the monitor and blood pressure mobile monitor. device are more than 5 meters away from each other. You selected Select the correct the wrong user profile on the profile on the blood pressure blood...
Emissions test Com Electromagnetic pli environment - ance guidance RF emissions Group 1 The device must emit CISPR 11 electromagnetic energy in order to perform its intended function. Nearby electronic equipment may be affected. RF emissions Class B CISPR 11 Harmonic emissions IEC applica...
Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey (a), should be less than the compliance level in each frequency range (b). Interference may occur in the vicinity of equipment marked with the following symbol: NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.