Screw Sensor into threaded hole of tab,
and fasten onto device being monitored.
If welding, weld must be a continuous bead. Tacking does not provide
sufficient acoustical coupling.
Do not arc weld on equipent connected to a SITRANS AS 100. Remove the
SITRANS AS 100 or disconnect electrically to avoid current flowing through
If gluing, use Loctite 326 adhesive, or equivalent. Follow manufacturer's
instructions to ensure proper adhesion.
Warning: Temperature at Sensor must not exceed minimum or
Ensure adequate isolation from hot
surfaces by providing additional
for standard Temperature
range model, maximum
temperature is 80 °C.
*maximum range dependent on model.
< 80 ºC
SITRANS AS 100 – INSTRUCTION MANUAL
Screw Sensor into disc, after welding or
bonding disc onto device being monitored.
Tab acts as heat sink.
Note: If the flange temperature is 100 °C
and the ambient temperature is 20 °C,
the Sensor temperature at the
electronics is below the maximum