Full certification In seca products you are purchasing not only technology matured over a century, but also quality confirmed by authorities, the legal system and by institutes. seca products comply with European directives, standards and national laws. With seca, you are buying the future.
In addition to the conventional determina- seca has been putting its experience at tion of weight, the seca 799 also has a the service of health for over 150 years function for determining body mass index.
– Close the cover and then turn the scale back up the right way. Note: the scale is prepared for use with a seca mains unit which is available as an accessory. The connection socket for the mains unit is in the base of the scale.
Fitting the column The column and display head are fitted to the weighing platform. – Guide the connecting cable through the opening. – Place the column in the recess. The display should be legible for people standing on the scale. –...
Switching the weighing range The seca 799 has two weighing ranges which can be selected. Weighing range 1 has a reduced maximum capacity but greater display resolution, whilst weighing range 2 allows you to use the maximum load capacity of the scale. Depending on the application in ques- tion, you decide whether you would prefer to use the fine display resolution or the high load range.
“HOLD“ appears in the digital display. The value is stored until the scale switches off automatically (operation on batteries). The triangle with the exclamation mark indicates that this frozen display is an additional facility which is not subject to calibration. Model 799...
– The HOLD function can be switched off by pressing the HOLD/TARE key again. appears in the display if the scale 0. 0 has no load on it, otherwise the current weight. The “HOLD“ display vanishes. Body Mass Index (BMI) Body Mass Index puts height and weight in relationship to one another, providing more accurate information than ideal Broca weight, for example.
The display will then stop flashing. – If there has been no activation of a function beforehand, remove the load from the scale and wait until is displayed, then weigh again. 0. 0 Model 799...
Recalibration is essential if one or more security marks are damaged. We recommend having your scale serviced by your local service agent before having it recalibrated. seca Customer Service will be pleased to give you further assistance. Changing the batteries When appear in the display, you should change the batteries.
Switchmode power adapter, 100-240V~ / 50-60Hz / 12V= / 0.5A seca no. 68-32-10-265 Attention! Use only seca original plug-in mains units with 9 V or controlled 12 V output voltage. Com- mercially-available mains units may supply a higher voltage than stated and thus lead to damage to the scale.