E 941/1 mobile unit
Mobile unit E 941/1, H 421, W 619, D 790 mm
with a magnetic strip for automatic mobile unit recognition
Please observe the Warning and Safety instructions for washer-disinfector
G 7827/28 or PG 8527/28.
If using automatic mobile unit recognition, please make sure the coding is
Areas of application:
Modules for laboratory glassware or for orthopaedic surgical instruments can be
placed in the E 941/1 mobile unit.
The modules have their own operating instructions.
Up to four modules can be placed in the mobile unit at once, depending on the
size of the items being processed.
Before loading the machine and before starting a programme, please check
– the sealing caps in the module connections are in good working order,
– the modules are correctly connected to the water intake in the mobile unit,
– the mobile unit is correctly connected to the water intake in the washer-
Placing modules into the mobile unit
^ Locate the module rollers on the runners, with the
spring-mounted water intakes facing the middle of
the mobile unit.
^ Push the module completely into the mobile unit.
Removing modules from the mobile unit
^ Lift the module up by the handles until the hooks clear the pins.
^ Then pull the module forwards and off the runners.
Miele Company Ltd., Fairacres, Marcham Road, Abingdon, Oxon, OX14 1TW, Tel 0330 160 6600,
Internet: http://www.miele.co.uk / ! 0297
Engage the hooks on the underside of the module
frame behind the pins on the runners on either side of
the mobile unit.
Make sure it clicks into position correctly.
Alteration rights reserved / Publication date 2015-11-01
When the module is correctly positioned, the water
intake on modules with integrated water connection
will connect correctly.
M.-Nr. 10 332 400 / 00