Mobile injector unit E 404/1 and E 405/1 for pipettes
Mobile injector unit, H 502, W 535, D 515 mm, E 405/1 with connection for hot air drying unit and holder for ML/2
magnetic strip (optional extra) for automatic mobile unit recognition (G 7835, G 7836, PG 8535 and PG 8536)
Use the LAB PIPETTES programme or increase the
water intake for the relevant reprocessing programmes
G 7836, PG 8536 + 5 litres
G 7835, G 7883, G 7883 CD, G 7893, PG 8535 + 6 litres
Placing pipettes into the injector unit:
^ With the pointed end facing downwards, place each
pipette into the appropriate holder in the base.
^ The upper ends of each pipette should be supported in
the relevant holders on the support rail.
Miele Company Ltd., Fairacres, Marcham Road, Abingdon, Oxon, OX14 1TW,
Tel 0845 365 0555, Fax 0845 365 0777, Internet: http://www.miele.co.uk /
Alteration rights reserved / Publication date 19.08.2010
This mobile injector unit can be used for reprocessing a
wide range of pipettes. Capacity:
max. length [mm] *
* The pipettes must not extend above the water intake pipe
or behond the sides of the mobile injector unit.
This mobile injector unit should only be used for processing
laboratory glassware which has been declared as suitable
for machine processing by the glassware manufacturer.
Please observe the Warning and Safety instructions
in the Operating Instructions for your washer-disinfector
as well as any instructions provided by the glassware
Broken glassware can cause serious injury.
All glassware should be properly inspected and tested
for cleanliness. Do not rely on a visual check alone.
^ Arrange pipettes with a volume of between 1 and 5 ml in
the injector sleeves in row c in the mobile injector unit.
The water pressure could push these pipettes out of the
funnel shaped holders in rows a and b.
10 to 100
10 to 25
1 to 10
M.-Nr. 07 913 000 / 00