Every time you fit a client, it’s an opportunity to build your reputation. When you choose ReSound, we understand the trust you’ve placed in us. We strive to provide you with flexible, simple-to-fit software so you can make a real dif-...
Airlink ReSound Aventa 3 supports two generations of Airlink; the Airlink dongle and the latest Airlink 2 . Both generations offer the great client experience of wireless fitting and the comfort it brings . Being the latest generation, Airlink 2 is the recommended option of the two .
Other programming interfaces Aventa 3 works with: • Airlink 2 • Airlink dongle • Speedlink • Hi-PRO • NOAHlink You can select programming interface on top of the Start Screen . Note: Animations in Connection Assistance will show you how to connect specific mod- els of hearing aids with the selected program- ming interface .
Start screen After launching the Aventa software you will see the Start screen . Select the patient’s experience with hear- ing aids for an optimized gain prescription . Let Aventa recommend the program de- faults or select them based on your patient interview .
Fit screen Gain Adjustment Adjust gain changes in increments of 1 dB, 2 dB or 3 dB . Choose fine-tune environmental programs and Wireless program . Muting and linking the hearing aids . Pair the hearing instruments with wireless accessories . Advanced Features Feature settings are accessed by clicking “Advanced Features”...
Summary screen Confirm number of programs . Customize the Clinician report and the Patient handout, which provides the patient with useful information . Type in notes and they will be printed on patient’s handout . Save options let you select whether to save data in both the hearing aids and database, only in one place, or not at all .
General Precautions • This Aventa fitting software will apply recommended amplification settings based on available audiometrical infor- mation . Settings will be specific to each fitting . • Manual modification of the fitting parameters will impact and change the amplification level prescribed . This change will be audible to the patient when the instruments are connected .
General Warnings • When connected, the Aventa Fitting Software controls the acoustic amplification levels in a hearing instrument . In some acoustic environments, over amplification can cause discomfort and damage to the patients hearing . • The Fitting System provides initial default amplification settings based on hearing threshold levels . These default settings could be higher than stable levels and cause feedback when amplification is first applied .
WARNING points out a situation that could lead to serious injuries, CAUTION indicates a situation that could lead to minor and moderate injuries . Any issues relating to the EU Medical Device Directive 93/42/EEC should be directed to ReSound A/S . WORLDWIDE HEADQUARTERS UNITED KINGDOM NEW ZEALAND...