ReSound LiNX 3D fitting guide A guide for professionals...
First Fitting with ReSound Smart Fit This fitting guide gives an overview of how to fit ReSound LiNX 3D wireless hearing instruments with ReSound Smart Fit. A ReSound LiNX 3D 961-DRW wireless binaural fitting is depicted. Not all features described in this fitting guide are applicable for ReSound LiNX 3D 7 and 5 fittings.
1. Select instrument configuration and click “Continue.” Both ReSound Aventa and ReSound Smart Fit are contained within one software installation. The Smart Launcher will detect both ReSound LiNX 3D and legacy instruments, and then retrieve the software appropriate for the connected device(s).
4. Confirm connected hearing instruments. Click “Continue” to proceed to the Patient Profile screen or or click “Accessories” to pair accessories. For further instruction, see the Pairing Accessories Fitting Screen Section in this guide. On the Fitting screen, adjustments can be made to gain, advanced features, and more. 1.
Directionality Binaural Directionality III uses ReSound’s 2.4 GHz wireless technology to allow the hearing DFS Ultra II instruments to work together, dynamically selecting the best microphone response for the listening This is the feedback control. It is activated when the hearing instruments have been calibrated.
NoiseTracker II NoiseTracker II uses spectral subtraction to reduce noise between words and syllables in many real-world Low Frequency Boost noisy environments. The Per Environment setting allows for personalized noise reduction settings in Should a patient with an Ultra Power (UP) hearing instrument state that the sound is not full enough or seven environments.
• Wired fittings: Ensure that the programming interface (Speedlink, Hi-PRO USB or NOAHlink) is connected to the fitting computer before launching ReSound Smart Fit. Launch ReSound Smart Fit. For follow-up fittings, the software that was used in the previous fitting will launch. 2. Once the searching pop-up...
Adjustments screen. 1. Click the “Go to Smart Fit” button in the left navigation panel to close ReSound Aventa and arrive at 1. Insert the battery, wait two minutes and close the battery door of each hearing instrument to the Smart Fit Launcher.
ReSound Smart Fit fitting software and the ReSound Smart 3D app. A ReSound LiNX 3D 961-DRW wireless binaural fitting is used as the example. Please note that not all features described in this guide are applicable to ReSound LiNX 3D 7 and 5 fittings. Getting Started Connect hearing instruments to the ReSound Smart Fit software.
Activate Consent Move Cloud Icon to Taskbar 1. Click “Start Consent.” The orange cloud icon should be visible in the taskbar so that the hearing care professional is notified when requests for assistance are received. If it is not already visible then: 1.
View Patient’s Assist Questionnaire There are two ways to view the Assist Questionnaire sent by the patient via the ReSound Smart 3D app: 1. Click the “Assist Questionnaire” box beneath the Data Logging box in the Sidebar. This will expand the questionnaire and enable it to be moved around the fitting screen during fine-tuning adjustments.
View Patient’s Preferred Settings Prepare to Send Package Click the “On” toggle for View Patient Settings at the bottom right of the Fitting screen. This will enable 1. After saving in the Fitting screen, a message will appear stating that the current fitting viewing of the patient’s preferred settings in the app.
4. Save the fitting. 5. Choose “Send Settings” on the Save screen as outlined in the previous section. The patient will receive the package in the ReSound Smart 3D app and the Remote Session will appear on the patient’s timeline.
The volume of the TSG is set using the volume slider. The volume should be set to a level that provides relief, but does not completely mask the tinnitus. (Figure 3) ReSound Smart Fit Note: The volume slider is in dBSPL. TSG fitting guide Figure 1: Below is a suggestion on how to first fit the TSG.