® in Noah and in the hearing instrument(s). Once the connection assumed here. Note that not all features mentioned here are available in ReSound Verso 7 and 5. process is finished, click “Continue” to proceed, or “Unite Accessories” to pair ReSound Unite™ accessories to the hearing instrument(s).
- even in the most difficult listening situations. It is available as a program option in dual-microphone ReSound Verso 9 BTEs as well as RIEs with wireless capabilities. Using Program Handling Toolbar ReSound’s 2.4 GHz wireless technology, the microphone modes between both hearing instruments are coordinated for an optimal binaural response.
Gain Adjustment Feature settings including Expansion, DFS Ultra™ II, Auto DFS, When fitting Verso hearing instruments, Aventa 3 offers a third input level curve for 65 dB SPL. This allows for greater NoiseTracker™ II, WindGuard , and directional options are flexibility for fine-tuning gains, as well as excellent precision for matching targets of generic prescriptive formulas.
These settings are available in all microphone-enabled programs. NoiseTracker™ II In ReSound Verso, NoiseTracker™ II is set to be environmentally dependent for all listening programs except Music. This means that the degree of noise reduction applied at any moment depends on the listening environment.
Adjust the Volume Control range Adjust beep levels and demo beeps Saving to database and instrument(s) in progress ReSound Unite™ Accessory Management Summary The functionality for managing accessories can be accessed both from the Start screen and from the Fit screen. In a new The summary page allows the clinician to print information for the patient, a third party, or clinic notes for a chart.