GE Oil & Gas Masoneilan * SVi 1000 Advanced Performance Digital Positioner Bench Quick Start (Rev B) Instruction Manual GE Data Classification: Public...
PROBLEMS ARISE WHICH ARE NOT COVERED SUFFICIENTLY FOR THE CUSTOMER/OPERATOR'S PURPOSES THE MATTER SHOULD BE REFERRED TO GE. THE RIGHTS, OBLIGATIONS AND LIABILITIES OF GE AND THE CUSTOMER/OPERATOR ARE STRICTLY LIMITED TO THOSE EXPRESSLY PROVIDED IN THE CONTRACT RELATING TO THE SUPPLY OF THE EQUIPMENT.
The goal of this procedure is to perform a configuration of the SVi1000 up to running Find Stops and Autotune on a bench. Additionally, this document also provides guidance on when to run signatures and produce associated reports. This procedure uses GE’s ValVue* 3 software (ver. 3.20 or later) and the SVi1000* DTM (ver. 2.20.0 or later).
Loop Calibrator (with wall power supply and loop test lead wires) and either a USB connector or a serial port connector for your laptop. The setup shown in Figure 1 is a GE product and can be purchased at: https://www.gemeasurement.com/test-calibration/calibrators/ups-iii-loop-calibrator-ma-v ®...
For security purposes, you are forced to change your password NOTE during the first use. GE recommends registering your software immediately. You are required to register after a 30 day trial period. After the trial period you won’t be able to use the software until you complete the registration process that the software guides you through.
GE recommends registering your software. NOTE GE recommends registering your software immediately. You are required to register after a 30 day trial period. After the trial period you won’t be able to use the software until you complete the registration process that the software guides you through.
1. Select the Full Wizard tab and ensure the mode is in Setup. Figure 9 Full Wizard Tab 2. Set the Current Slot to either 0 for ATO or 8 for ATC. This procedure moves the valve. Procedures (e.g. Find Stops, Auto Tune, Step Test and Ramp Test) should NOT be invoked if the ValVue sequencer is running.
b. Clicking the Disconnect icon on the ValVue 3 toolbar. Clicking Network > Exit to close the DTM. If you have unsaved data, click the Download Parameters to Device icon to save. 3. Connect SVi1000 wires to next item on order. Click Find New Devices. Go to step 10 of “2. Software Install and Configuration”...
Appendix A: Download Instructions Masoneilan Software 1. Go to the Download Center (http://www.ge-mcs.com/en/download.html), make selections as shown (Figure 11) and click Search. Figure 11 Download Center: Search for Valve 3 The results appear (Figure 12). Figure 12 Search Results: ValVue 3 Figure 13 Download Center: Search for SVi1000 DTM 2.
3. Click I Accept for the terms (Figure 14). Figure 14 Software Terms 4. Fill in the Registration Form and click Submit (Figure 15). Filling in this form registers the download. It does not register the NOTE software. Download, install and follow registration instructions in software help.
Fax: 88+91-11-5-1659635 Fax: 88+65-6861-7172 Fax: 88+1-281-671-1735 Jacksonville, Florida Phone: +1-844-VALVE-GE www.geoilandgas.com/valves *Denotes a registered trademark of the General Electric Company. Other company names and product names used in this document are the registered trademarks or trademarks of their respective owners.